A standard curve based method for relative real time PCR data processing
 Alexey Larionov^{1}Email author,
 Andreas Krause^{2} and
 William Miller^{3}
DOI: 10.1186/14712105662
© Larionov et al; licensee BioMed Central Ltd. 2005
Received: 11 November 2004
Accepted: 21 March 2005
Published: 21 March 2005
Abstract
Background
Currently real time PCR is the most precise method by which to measure gene expression. The method generates a large amount of raw numerical data and processing may notably influence final results. The data processing is based either on standard curves or on PCR efficiency assessment. At the moment, the PCR efficiency approach is preferred in relative PCR whilst the standard curve is often used for absolute PCR. However, there are no barriers to employ standard curves for relative PCR. This article provides an implementation of the standard curve method and discusses its advantages and limitations in relative real time PCR.
Results
We designed a procedure for data processing in relative real time PCR. The procedure completely avoids PCR efficiency assessment, minimizes operator involvement and provides a statistical assessment of intraassay variation.
The procedure includes the following steps. (I) Noise is filtered from raw fluorescence readings by smoothing, baseline subtraction and amplitude normalization. (II) The optimal threshold is selected automatically from regression parameters of the standard curve. (III) Crossing points (CPs) are derived directly from coordinates of points where the threshold line crosses fluorescence plots obtained after the noise filtering. (IV) The means and their variances are calculated for CPs in PCR replicas. (V) The final results are derived from the CPs' means. The CPs' variances are traced to results by the law of error propagation.
A detailed description and analysis of this data processing is provided. The limitations associated with the use of parametric statistical methods and amplitude normalization are specifically analyzed and found fit to the routine laboratory practice. Different options are discussed for aggregation of data obtained from multiple reference genes.
Conclusion
A standard curve based procedure for PCR data processing has been compiled and validated. It illustrates that standard curve design remains a reliable and simple alternative to the PCRefficiency based calculations in relative real time PCR.
Background
Data processing can seriously affect interpretation of real time PCR results. In the absence of commonly accepted reference procedures the choice of data processing is currently at the researcher's discretion. Many different options for data processing are available in software supplied with different cyclers and in different publications [1–7]. However, the basic choice in relative real time PCR calculations is between standard curve and PCRefficiency based methods. Compared to the growing number of studies addressing PCR efficiency calculations [3, 5, 8–10] there is a shortage of publications discussing practical details of the standard curve method [11]. As a result, the PCR efficiency approach appears as the method of choice in data processing for relative PCR [12]. However, when reliability of results prevails over costs and labor load, the standard curve approach may have advantages.
The standard curve method simplifies calculations and avoids practical and theoretical problems currently associated with PCR efficiency assessment. Widely used in many laboratory techniques this approach is simple and reliable. Moreover, at the price of a standard curve on each PCR plate it also provides the routine validation for methodology. To benefit from the advantages of the standard curve approach and to evaluate its practical limitations we designed a data processing procedure implementing this approach and validated it for relative real time PCR.
Results
Description of the data processing procedure
Source data
Raw fluorescence readings were exported from Opticon Monitor software and processed in MS Excel using a VBA script (the mathematical formulae, script and samples of source data are attached to the electronic version of publication, see Additional files 1 and 2).
Noise filtering
Crossing points calculation
Standard curve calculation
A standard curve was derived from the serial dilutions by a customary way. Relative concentrations were expressed in arbitrary units. Logarithms (base 10) of concentrations were plotted against crossing points. Least square fit was used as the standard curve.
Threshold selection
The optimal threshold was chosen automatically. The VBA script examined different threshold positions calculating coefficient of determination (r^{2}) for each resulting standard curve. The maximum coefficient of determination pointed to the optimal threshold (typically the maximum r^{2} was larger than 99%).
Calculating means and variances of means for crossing points in PCR replicas
The optimal threshold was used to calculate CPs for unknown samples. Means and variances of means were then calculated for CPs in PCR replicas.
Derivation of nonnormalized values from crossing points
The nonnormalized values were calculated from the CPs' means by the standard curve equation followed by exponent (base 10). The variances were traced by the law of error propagation.
Summarizing data from several reference genes to a single normalizing factor

Arithmetic mean (deprecated),

Geometric mean (recommended).
Calculation of normalized results for target genes
The final results representing relative expression of target genes were calculated by dividing the nonnormalized values by the above normalization factor. The normalized results' variances were derived by the law of error propagation.
When confidence intervals or coefficients of variation were needed they have been calculated from the corresponding variances (see Additional file 1 with formulae for details).
Procedure testing and validation
Primers' sequences
Short name  Full name  GenBank number  Primers 

SCGB2A2  Mammaglobin 1 (Secretoglobin, family 2A, member 2)  NM_002411  TCC AAG ACA ATC AAT CCA CAA G AAA ATA AAT CAC AAA GAC TGC TG 
SCGB2A1  Mammaglobin 2 (Secretoglobin, family 2A, member 1)  NM_002407  AAG ACC ATC AAT TCC GAC ATA CAC CAA ATG CTG TCG TAC ACT 
CCNB1  Cyclin B1  NM_031966  CAT GGT GCA CTT TCC TCC TT CAG GTG CTG CAT AAC TGG AA 
CKS2  CDC28 protein kinase regulatory subunit 2  NM_001827  TTC ATG AGC CAG AAC CAC AT CTC GTG CAC AGG TAT GGA TG 
PTN  Pleiotrophin (heparin binding growth factor 8, neurite growthpromoting factor 1)  NM_002825  GTG CAA GCA AAC CAT GAA GA GCT CGC TTC AGA CTT CCA GT 
LPIN2  Lipin 2  NM_014646  TTG TTG CTG CAG ATT GAT CC CCA AAT GGC AAT GGA TTT TC 
ACTB  Actin, beta  NM_001101  GGA GCA ATG ATC TTG ATC TT CCT TCC TGG GCA TGG AGT CCT 
GAPD  glyceraldehyde3phosphate dehydrogenase  NM_002046  TGC ACC ACC AAC TGC TTA GC GGC ATG GAC TGT GGT CAT GAG 
Crossing points' distributions observed in PCR replicas
Plate  Number of replicates  Mean CP  SD  CV  Skewness  Kurtosis  KolmogorovSmirnov test 

1  96  21.48  0.06  0.3%  0.1  0.1  Normal 
2  94  18.09  0.07  0.4%  1.5  5.7  Sharper than normal 
3  96  20.09  0.04  0.2%  0.1  0.3  Normal 
4  96  18.13  0.10  0.5%  0.5  1.0  Normal 
Discussion
PCR data processing is a complex procedure that includes a number of steps complementing each other. Many different options have been suggested by different authors at each step of PCR data processing. In the discussion below we go through our procedure on a steptostep basis shortly discussing the available options and explaining our choices. In general, we preferred the simplest functioning solutions. In statistical treatment we looked for valid practical estimations rather than for mathematically exact solutions. Because of lack of relevant theoretical data we paid especial attention to the amplitude normalisation and to statistical processing of intraassay PCR replicas. To validate these sections of our procedure we had to address some basic theoretical issues.
PCR data processing may need to be optimized for specific PCR machines and chemistry. The discussed processing was optimized for data obtained on an Opticon Monitor 2 machine (MJ Research) using the QuantiTect SYBR Green PCR kit (Qiagen).
Smoothing
Smoothing is necessary if noticeable nonspecific scattering from cycle to cycle is observed on the raw fluorescence plots. Apart from moving averages there are other more sophisticated mathematical approaches to filter this kind of noise e.g. sigmoidal fitting [13]. However, this fit is no more than a mathematical abstraction fitting PCR plot. Until the development of a genuine mathematical model of real time PCR, all other fits will not be related to PCR per se. Therefore, since simple 3 point moving average produced acceptable results there was no obvious need for more complex methods.
Background subtraction
Background subtraction is a common step in PCR data processing. Often it requires operator's involvement to choose between several available options (e.g. subtraction of minimal value through the run, subtraction of average over a certain cycle ranges, different kinds of "trends", etc). To avoid the operator involvement we always subtract the minimal value observed in the run. This option has a clear interpretation and works well. It is important that the baseline subtraction is performed after smoothing. So the noise potentially affecting minimal values has already been reduced before baseline subtraction.
Amplitude normalization
Amplitude normalization unifies plateau positions in different samples. Although amplitude normalization was available in some versions of LightCycler software and has been used by some researchers [14] this step still is not common in PCR data processing. The caution with regard to the amplitude normalization is probably caused by current lack of understanding of the plateau phase in PCR.
Amplitude normalization is based on the suggestion that in ideal PCR, output is determined by the initially available PCR resources. In this case PCRs prepared from the same master mix will run out of the same limiting resource in different samples. The resource can run out sooner (abundant template) or later (rare template) but finally the same amount of PCR products will be produced in all samples. This assumption is valid for ideal PCR but in practice it may not always hold (for example, nonspecific PCR products may also consume PCR resources). The factors potentially leading PCR to the plateau include utilization of primers or nucleotides, thermal inactivation of DNA polymerase, competition between primers and PCR products for annealing, enzyme inactivation by PCR products and accumulation of inhibitors [15]. The plateau may also be affected by factors influencing the detection of PCR products: e.g. by PCR volume and by concentration of probe or SYBRGreen in PCR mix [14, 16, 17]. In practice the plateau phase is probably caused by different factors depending on the particular PCR design and PCR mix composition.
In this work we used QuantiTect SYBR Green PCR kit (Qiagen). With this kit neither increase of primers nor addition of enzyme notably affected the plateau positions (Figure 7). The fact that bands on PCR gels were remarkably enlarged by additional staining with SYBR Green (Figure 6) suggests that the plateaus observed in PCRs could had been caused simply by limited SYBR Green concentration. Therefore, in samples prepared with the same master mix, the plateau scattering could be considered as a nonspecific noise and should be removed.
Threshold selection
As long as the standard curve provides both basis and empirical validation for PCR results the threshold may be put at any level where it produces a satisfactory standard curve. At the same time, the linearity of standard curve is theoretically explained at exponential phase of PCR only. Therefore, the common practice is to put the threshold as low as possible to cross the fluorescence plots in the exponential phase. For this reason we usually restrict the search of the optimal threshold position to the lower half of the fluorescence plot.
Crossing point calculation
Currently the most established methods of crossing point calculations are the fit point method and the second derivative maximum method [4]. The fit point method reliably allocates the threshold level in the exponential phase and reduces minor inaccuracies by aggregating data from several points. The second derivative maximum method eliminates interactivity during threshold selection and baseline subtraction. These are robust and reliable methods.
Our calculation method also produces good results. In addition, it is simple and does not alter the initial mathematical definition of crossing points.
Statistical treatment of PCR replicas
The next step in the data processing is derivation of results from crossing points. Two separate issues need to be addressed during this step: (i) bestfit values and (ii) errors in replicates. Calculation of bestfit values is simple with standard curve methodology (see formulae in Additional file 1) but statistical assessment of errors in replicates requires detailed consideration.
Description and interpretation of intraassay PCR variation
PCR uncertainty is usually characterized by coefficient of variation. This reflects the fact that the errors propagated to nonnormalized values and to final results are higher at higher bestfit values. This is not always the case with the crossing points. However, coefficients of variation still may be used for rough comparison of CPs' dispersions because the CPs' absolute values vary in quite a limited range (typically between 20 and 30 cycles).
Importantly, that during PCR interpretation the statistical significance of differences between samples should not be based on intraassay variation. IntraPCR replicates account only for errors originated from PCR. At the same time the uncertainty in final results is usually more affected by prePCR steps [1]. In this case the replicates of the whole experiment (including sampling, RNA extraction and reverse transcription) are needed to derive statistical differences between samples. If the amount of starting material is limited or replicates are unavailable (for example when studying tumor biopsies) the preliminary assessment of replicates in an experimental set of similar samples is required to base statistical comparison between samples (type B evaluation of uncertainty according to Taylor and Kuyatt [19]). This type of statistical treatment is not included in the described data processing. Even though in our experiments the intraassay PCR variation can not be directly used for statistical inferences, we routinely use it as an internal quality check for PCR.
Starting point for statistical assessment
Two different approaches may be utilized for initial statistical handling of intraassay PCR replicates. Either CP values are first averaged and then transformed to nonnormalized values or vice versa. Both approaches may yield similar results, as long as the arithmetic mean is used for the CP values and geometric mean for the nonnormalized quantities. We prefer to start statistical assessment using unmodified source data i.e. we average crossing points before transformation to the nonnormalized values.
Crossing point distribution in PCR replicas
To choose appropriate statistical methods to deal with crossing points, we started from the assessment of crossing points' distributions in PCR replicates. Distributions of crossing points were studied in four PCR plates each of those represented a 96× replicate. The distributions were close to the Normal (Table 2, Figure 4). Combined analysis of a number of PCR reactions, made in triplicates or quadruplicates, confirmed this result (data not shown). Therefore, Normal distribution satisfactorily reflects the distribution of crossing points in PCR replicates. This allowed us to use arithmetic mean and mean's variances to estimate bestfit values and their uncertainty in crossing points.
Error propagation
Magnitude of propagated error at different steps of data processing
SD in crossing points  CV in crossing points  CV in nonnormalized values  CV in normalized results 

0.1  0.5%  7%  10% 
0.2  1.0%  14%  20% 
0.3  1.5%  22%  31% 
0.4  2.0%  28%  40% 
0.6  3.0%  45%  66% 
Additionally the analysis confirmed the remarkable increase of relative variation at each step of data processing. E.g. 2% CV at crossing points resulted to 28% CV in the nonnormalized values and to 40% CV in the final results (Table 3). This also complicates interpretation of results with high dispersion in crossing points.
Standard curves
In line with the common practice, we interpreted the standard curve as an ordinary linear function ignoring its statistical nature and uncertainty because the uncertainty was usually quite small (typical coefficient of determination above 99%). With sufficient number and range of standard dilutions and proper laboratory practice it is always should be possible to produce the standard curve of sufficient quality.
Specific design of standard curves may differ for different genes depending on the variability of their expression. For relatively stabile genes (e.g. Actin beta or GAPD) we usually were able to obtain good standard curves using 5–6 twofold dilutions. To cover the dynamic range for genes with less stable expression (e.g. Mammaglobin 1 in breast cancers) more dilutions (up to 8) and/or higher factor at each dilution (3–5 fold) were needed. We usually run standards in triplicates (as well as the target specimens).
Even though the standard curves could be quite reproducible [12] we consider the presence of standard curves on each plate to be a good laboratory practice. Additionally, there is no great economy in sharing standard curves between PCR plates, when the plates are filled up with samples. For example, 6point standard curve in triplicates takes just 18 cells: this is less than 20% of 96plate and less than 5% of 386plate. Therefore sharing of standard curves reduces costs and labour only in pilot experiments with small number of samples. However, even in pilot experiments the repeatability of shared standard curves should be validated on a regular basis.
Summarizing data from several reference genes
Several reference genes are required for accurate relative quantification [1, 20]. Different ways may be used to derive a single normalizing factor out of several genes. To explore this in the attached version of VBA script we made available two options: arithmetic and geometric mean.
Arithmetic mean is the most "intuitive" way. However, it has a major disadvantage: it depends on arbitrary choice of the absolute values for reference genes. For example, the normalizing factor will differ, if a reference gene is described either as a fraction of 1 (absolute values from 0 to 1) or in percents (values 0% to 100%). Importantly, this can change the relative values of the normalizing factor in different samples. In contrast, if geometric mean is used, the arbitrary choice of units for any reference gene will not affect the relative values of normalizing factor in different samples. Neither arithmetic nor geometric mean accounts for differences in uncertainties of different reference genes. In practice this implies similar variances in all reference genes. This assumption seems reasonable in most of the cases. However, if this assumption does not hold the weights reciprocal to variances could be introduced.
Obviously, the different ways of summarizing data from reference genes will produce different results. At the same time, at truly stable expression of reference genes the general tendencies in results should be similar. Currently we calculate the single normalizing factor by geometric mean, because it better fits to the relative nature of measurements as well as to the logarithmic scale of gene expression changes [20, 21].
Unfortunately common practice tends to ignore the uncertainty of normalizing factor. Our procedure estimates this uncertainty using the law of error propagation (see formulae in Additional file 1).
Methods based on PCR efficiency and individual shapes of fluorescent plots
Standard curve approach was chosen for our procedure because currently PCR efficiency assessment may complicate data processing. The main complication is that actual efficiency of replication is not constant through the PCR run being high at exponential phase and gradually declining toward the plateau phase. However, most current methods of PCR efficiency assessment report "overall" efficiency as a single value. Additionally, PCR efficiency may be calculated in different ways that can "overestimate" or "underestimate" the "true" PCR efficiency [12]. In contrast, the standard curve method is based on a simple approximation of data obtained in standard dilutions to unknown samples.
At present the most popular method of PCR efficiency assessment is based on the slope of standard curve. This method does not account for PCR efficiencies in individual target samples. In contrast, recent publications on PCR efficiency assessment were concentrated on the analysis of individual shapes of fluorescence plots [8–10]. Potentially this may lead to better mathematical understanding of PCR dynamic and to new practical solutions in PCR quantification [13].
Limitations of our data processing
This section summarizes conditions that must be adhered to in order to obtain valid results with our data processing:

all PCRs must achieve doubtless plateau and no nonspecific PCR products should be observed to use amplitude normalization;

standard curves with coefficient of determination above 99% are required to ignore uncertainty of regression and to validate the use of amplitude normalization;

low dispersion in PCR replicates (crossing points' CV < 1% or SD < 0.2) is required to use the conventional statistical methods.
These limitations are linked: amplitude normalization provides the low dispersion in replicas needed for statistical treatment.
Conclusion
In this article we described a procedure for relative real time PCR data processing. The procedure is based on the standard curve approach, does not require PCR efficiency assessment, can be performed in fully automatic mode and provides statistical assessment of intraassay PCR variation. The procedure has been carefully analyzed and tested. The standard curve approach was found a reliable and simple alternative to the PCRefficiency based calculations in relative real time PCR.
Methods
Tissue samples, RNA extraction, reverse transcription
Breast cancer biopsies were taken from 21 patients before and after treatment with an aromatase inhibitor. Samples were obtained in the Edinburgh Breast Unit (Western General Hospital, Edinburgh) with patients' informed consent and ethical committee approval. Biopsies were snap frozen and stored in liquid nitrogen until RNA extraction. Before RNA extraction the frozen tissue was defrosted and stabilized in RNAlaterICE reagent (Ambion). Total RNA was extracted with RNeasymini columns (Qiagen). Amount and purity of RNA were evaluated by spectrophotometer. RNA integrity was confirmed by agarose gel electrophoresis.
 1)
oligo(dT)_{20} primers and dNTPs were added to total RNA,
 2)
the mix was heated to 65°C for 5 min and then chilled on ice,
 3)
firstStrand buffer, DDT, RNase inhibitor (RNaseOUT, Invitrogen) and Reverse transcriptase were added to specimens,
 4)
reverse transcription was carried out for 60 minutes at 50°C.
PCR
 1.
Aliquots of cDNA samples running on the same plate were pooled and the pool was used as calibrator.
 2.
cDNAs were diluted with water prior PCR.
 3.
The set of samples consisting of the diluted cDNAs and the dilutions of the calibrator were used for several PCR plates: one plate for each gene.
 4.
For each sample the whole PCR mix including primers and cDNA was prepared before dispensing into the plate.
 5.
Samples were loaded to 96× PCR plates by 15 μl per cell in triplicates or quadruplicates.
Primer's sequences are given in Table 1. Primers were designed basing on the sequences published in GenBank and using Primer3 software [22]. To avoid genomic DNA amplification the primers were either located in different exons or across exonexon boundaries. Primers were synthesized in Sigma Genosys or in Cancer Research UK. PCR was performed using QuantiTect SYBR Green PCR kit (Qiagen), Opticon2 PCR machine (MJ Research), white 96× PCR plates and plain PCR caps (MJ Research). The cycling parameters for all genes were the following: hotstart 95°C 15 min, 45 cycles of (denaturation 94°C 15 sec, annealing 56°C 30 sec, elongation 72°C 30 sec, plate read), final elongation 72°C 5 min, melting curve 65–95°C. Gradient PCRs confirmed 56°C as appropriate annealing temperature for all primers.
Several additional PCRs were run with different amount of primers (0.1 μM, 0.3 μM, 0.9 μM), different amount of enzyme (0.8U, 1.5U and 3.1U of HotStarTaq, Qiagen were added to 15 μl PCRs made with QuantiTect SYBR Green PCR mix, Qiagen) and different caps (domed and plain caps, MJ Research).
PCR product electrophoresis
Electrophoreses were run immediately after PCRs. 10 μl of PCR products were mixed with 2 μl of loading buffer. 6 μl of the mix per well was loaded into 10% PAAG (TBE Ready Gel, Biorad). Electrophoresis was run at 100 V for ~1 hr using MiniProteanII cell (Biorad).
Prior electrophoresis 1 μl of 1:100 SybrGreen1 (Molecular Probes) was added into molecular weight marker (PCR Low Ladder Set, Sigma) but not into the PCR samples. After electrophoresis the gels were stained for 10 min in fresh prepared 1:10000 SybrGreen1 (Molecular Probes). Photos were taken before and after staining using the GelDocMega4 gel documentation system (Uvitec).
96× PCR replicas
To study distributions of crossing points in PCR replicas four PCR plates have been run with a 96× replica on each. The distributions were evaluated using histograms, skewness and kurtosis measures, and the KolmogorovSmirnov test for Normality (see Table 2 and Figure 4).
Normal distribution transformation through the data processing
Basing on the above empirical observations (Table 2, Figure 4) the crossing points were simulated by sampling from the Normal. Samples of 1,000 random normal numbers were obtained using standard Excel data analysis tool. A pair of such samples was used to simulate CPs for one target and one reference genes. Then the simulated CPs were processed in the same way as real PCR data. The distributions obtained at each step of data processing were evaluated for normality by histograms, skewness and kurtosis measures, and the KolmogorovSmirnov test.
Parameters used in calculations were close to actual parameters typically observed in our PCRs (MeanCP = 20, Slope = 0.3, Intercept = 7). The resulted true values for nonnormalized and normalized results were 10 and 1 correspondingly.
To study error propagation at different initial dispersions we performed simulations using the Normal distributions with different variances (CV 0.5%, 1%, 1.5%, 2%, 3%, 4% and 5%; the means were always 20). Detailed illustration for CV 1% is presented in Figure 12. The summary of simulation results is presented in Figure 5 and Table 3.
Excel VBA macros
The calculations where performed using MS Excel VBA script included to the electronic version of publication (see Additional file 2).
List of abbreviations
 GAPD :

glyceraldehyde3phosphate dehydrogenase
 CP (CPs):

crossing point (crossing points)
 SD:

standard deviation
 CV:

coefficient of variation
 r^{2}:

coefficient of determination in linear regression
Declarations
Authors’ Affiliations
References
 Bustin SA: Quantification of mRNA using realtime reverse transcription PCR (RTPCR): trends and problems. J Mol Endocrinol 2002, 29: 23–39. 10.1677/jme.0.0290023View ArticlePubMed
 Muller PY, Janovjak H, Miserez AR, Dobbie Z: Processing of gene expression data generated by quantitative realtime RTPCR. Biotechniques 2002, 32: 1372–4, 1376, 1378–9.PubMed
 Pfaffl MW: A new mathematical model for relative quantification in realtime RTPCR. Nucleic Acids Res 2001, 29: e45. 10.1093/nar/29.9.e45PubMed CentralView ArticlePubMed
 Pfaffl MW, Horgan GW, Dempfle L: Relative expression software tool (REST) for groupwise comparison and statistical analysis of relative expression results in realtime PCR. Nucleic Acids Res 2002, 30: e36. 10.1093/nar/30.9.e36PubMed CentralView ArticlePubMed
 Livak KJ, Schmittgen TD: Analysis of relative gene expression data using realtime quantitative PCR and the 2(Delta Delta C(T)) Method. Methods 2001, 25: 402–408. 10.1006/meth.2001.1262View ArticlePubMed
 Roshe Applied Science: Overview of LightCycler Quantification Methods. Technical Note No LC 10 2003.
 Applied Biosystems: Guide to Performing Relative Quantitation of Gene Expression Using RealTime Quantitative PCR. 2004.
 Tichopad A, Dilger M, Schwarz G, Pfaffl MW: Standardized determination of realtime PCR efficiency from a single reaction setup. Nucleic Acids Res 2003, 31: e122. 10.1093/nar/gng122PubMed CentralView ArticlePubMed
 Liu W, Saint DA: A new quantitative method of real time reverse transcription polymerase chain reaction assay based on simulation of polymerase chain reaction kinetics. Anal Biochem 2002, 302: 52–59. 10.1006/abio.2001.5530View ArticlePubMed
 Bar T, Stahlberg A, Muszta A, Kubista M: Kinetic Outlier Detection (KOD) in realtime PCR. Nucleic Acids Res 2003, 31: e105. 10.1093/nar/gng106PubMed CentralView ArticlePubMed
 Rutledge RG, Cote C: Mathematics of quantitative kinetic PCR and the application of standard curves. Nucleic Acids Res 2003, 31: e93. 10.1093/nar/gng093PubMed CentralView ArticlePubMed
 Pfaffl MW: Quantification strategies in real time PCR. IUL biotechnology series ; 5. In AZ of quantitative PCR. Edited by: Bustin SA. La Jolla, CA, International University Line; 2004.
 Rutledge RG: Sigmoidal curvefitting redefines quantitative realtime PCR with the prospective of developing automated highthroughput applications. Nucleic Acids Res 2004, 32: e178. 10.1093/nar/gnh177PubMed CentralView ArticlePubMed
 Wittwer CT, Herrmann MG, Moss AA, Rasmussen RP: Continuous fluorescence monitoring of rapid cycle DNA amplification. Biotechniques 1997, 22: 130–1, 134–8.PubMed
 Kainz P: The PCR plateau phase  towards an understanding of its limitations. Biochim Biophys Acta 2000, 1494: 23–27.View ArticlePubMed
 Zipper H, Lämmle K, Buta C, Brunner H, Bernhagen J, Vitzthum F: Investigations on the binding of SYBR Green I to doublestranded (ds)DNA: In Proceedings of the joint annual fall meeting , German Society for Biochemistry and Molecular Biology (GBM) & German Society for Expermental and Clinical Pharmacology and Toxicology (DGPT) September 7–10 2002; Halle (Saale), Germany. ; 2002:177.
 Vitzthum F, Geiger G, Bisswanger H, Brunner H, Bernhagen J: A quantitative fluorescencebased microplate assay for the determination of doublestranded DNA using SYBR Green I and a standard ultraviolet transilluminator gel imaging system. Anal Biochem 1999, 276: 59–64. 10.1006/abio.1999.4298View ArticlePubMed
 Freshwater S, van der Valk A, O'Shaughnessy M, Ng S, Baker S: The effect of consumable type on the sensitivity and reproducibility of qPCR: In Proceedings of the 1^{ st } International qPCR Symposium and Application Workshop 3rt  6th March 2004; FreisingWeihenstephan, Germany. Edited by: Pfaffl MW. ; 2004:88.
 Taylor BN, Kuyatt CE: Guidelines for evaluating and expressing the uncertainty of NIST measurement results. In NIST technical note ; 1297. 1994 edition. Gaithersburg, MD, U.S. Department of Commerce, Technology Administration, National Institute of Standards and Technology; 1994:20 p..
 Vandesompele J, De Preter K, Pattyn F, Poppe B, Van Roy N, De Paepe A, Speleman F: Accurate normalization of realtime quantitative RTPCR data by geometric averaging of multiple internal control genes. Genome Biol 2002, 3: RESEARCH0034. 10.1186/gb200237research0034PubMed CentralView ArticlePubMed
 Szabo A, Perou CM, Karaca M, Perreard L, Quackenbush JF, Bernard PS: Statistical modeling for selecting housekeeper genes. Genome Biol 2004, 5: R59. 10.1186/gb200458r59PubMed CentralView ArticlePubMed
 Rozen S, Skaletsky HJ: Primer3 on the WWW for general users and for biologist programmers. In Bioinformatics Methods and Protocols: Methods in Molecular Biology. Edited by: Krawetz S and Misener S. Totowa, NJ,, Humana Press; 2000:365–386.
 Larionov AA, Hulme MJ, Miller WR: Amplitude normalization in real time PCR data processing: 3rt  6th March 2004; FreisingWeihenstephan, Germany. Edited by: Pfaffl MW. ; 2004:56–57.
 Larionov AA, Miller WR: Data processing in real time PCR: In Proceedings of the 1^{ st } International qPCR Symposium and Application workshop FreisingWeihenstephan, Germany. Edited by: Pfaffl MW. ; 2004:28–29.
Copyright
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.