Improved classification accuracy in 1 and 2dimensional NMR metabolomics data using the variance stabilising generalised logarithm transformation
 Helen M Parsons^{1},
 Christian Ludwig^{2},
 Ulrich L Günther†^{2} and
 Mark R Viant†^{1, 3}Email author
DOI: 10.1186/147121058234
© Parsons et al; licensee BioMed Central Ltd. 2007
Received: 06 March 2007
Accepted: 02 July 2007
Published: 02 July 2007
Abstract
Background
Classifying nuclear magnetic resonance (NMR) spectra is a crucial step in many metabolomics experiments. Since several multivariate classification techniques depend upon the variance of the data, it is important to first minimise any contribution from unwanted technical variance arising from sample preparation and analytical measurements, and thereby maximise any contribution from wanted biological variance between different classes. The generalised logarithm (glog) transform was developed to stabilise the variance in DNA microarray datasets, but has rarely been applied to metabolomics data. In particular, it has not been rigorously evaluated against other scaling techniques used in metabolomics, nor tested on all forms of NMR spectra including 1dimensional (1D) ^{1}H, projections of 2D ^{1}H, ^{1}H Jresolved (pJRES), and intact 2D Jresolved (JRES).
Results
Here, the effects of the glog transform are compared against two commonly used variance stabilising techniques, autoscaling and Pareto scaling, as well as unscaled data. The four methods are evaluated in terms of the effects on the variance of NMR metabolomics data and on the classification accuracy following multivariate analysis, the latter achieved using principal component analysis followed by linear discriminant analysis. For two of three datasets analysed, classification accuracies were highest following glog transformation: 100% accuracy for discriminating 1D NMR spectra of hypoxic and normoxic invertebrate muscle, and 100% accuracy for discriminating 2D JRES spectra of fish livers sampled from two rivers. For the third dataset, pJRES spectra of urine from two breeds of dog, the glog transform and autoscaling achieved equal highest accuracies. Additionally we extended the glog algorithm to effectively suppress noise, which proved critical for the analysis of 2D JRES spectra.
Conclusion
We have demonstrated that the glog and extended glog transforms stabilise the technical variance in NMR metabolomics datasets. This significantly improves the discrimination between sample classes and has resulted in higher classification accuracies compared to unscaled, autoscaled or Pareto scaled data. Additionally we have confirmed the broad applicability of the glog approach using three disparate datasets from different biological samples using 1D NMR spectra, 1D projections of 2D JRES spectra, and intact 2D JRES spectra.
Background
Metabolomics relies extensively upon the multivariate analysis of data [1]. Unsupervised data mining tools such as principal component analysis (PCA) [2, 3] and hierarchical clustering [4, 5] and supervised methods such as partial least squares discriminant analysis (PLSDA) [1, 6] are commonly used to search for patterns and other features within metabolomic data sets. Many multivariate techniques evaluate possible relationships between samples by examining the variance of the data [3, 7], the most notable being PCA for which principal components (PCs) are calculated along the directions of maximum variance. Hence the structure of the variance within a metabolomics data set can have a major effect on the output of the multivariate analyses. It is therefore important to assess (and modify appropriately) the variance structure of a metabolomics data set prior to multivariate analysis. Variation between samples can be broadly classified into one of two types – 'technical' and 'biological' [2, 8]. Technical variance is created by the experimental procedure, and includes sample preparation and analytical measurement errors, whilst biological variance is the inherent variation between samples created by genetic differences, pathological or environmental factors, etc [8]. Clearly, the technical variance does not contribute any useful information to discriminate between different biological sample classes and so, ideally, this variance would not contribute to any multivariate analyses.
Autoscaling is a processing technique in which the variance of each variable is scaled to unity and the mean of each variable is set to zero [5]. In Pareto scaling each variable's intensity is scaled by the square root of the standard deviation of that variable [3], producing a data set where the variance changes from variable to variable, but the range of variance across each spectrum is much reduced from the initial, unscaled data.
The glog is a transformation that was originally applied to microarray data [12, 13] and is based on the twocomponent error model [14]. Specifically, the measurement error of an observation is characterised by one component representing the error of the data as being proportional to the intensity of the measurement, and a second, additive, component of the error characterising the noise. Previously, the glog transform has been applied to onedimensional (1D) NMR data as first shown by Purohit et al [11]. Unlike autoscaling and Pareto scaling, the glog transform initially requires a single parameter to be calibrated from a series of 'technical replicates'. These replicates must be recorded from one biological sample that has been divided into five or more components, each of which is subject to independent sample preparation and NMR analysis. The variance within this data set arises solely from technical sources [2, 8], upon which the glog transform is calibrated [11]. Hence, when the glog is then applied to a biological data set it effectively reduces the amount of technical variance present, leaving the biological variance to dominate any subsequent multivariate analysis. To date, the glog transform has not been compared against other processing techniques used in metabolomics. Furthermore, the calibrated glog transform has only been tested using a single, relatively small 1D NMR data set [11]. Recently, due largely to severe peak congestion in 1D NMR spectra, there has been a significant increase in the range of 2D NMR experiments conducted in metabolomics [15–21]. Although many of these experiments require substantially longer acquisition times and so are not appropriate for high throughput metabolomics, 2D Jresolved (JRES) spectroscopy has been shown to provide spectra with low peak congestion and high metabolite specificity in a short acquisition time [15, 20]. Consequently, several multivariate analyses of 1D projections of 2D JRES spectra have been reported [15, 18–21]. To our knowledge the applicability of the glog transform, including the initial calibration of the function using technical replicates, has not been evaluated for these 1D projections of 2D JRES spectra, nor for the analysis of the intact 2D JRES spectra.
Here, we first aimed to evaluate comprehensively the glog transform compared to two other commonly used scaling methods in NMR metabolomics as well as against unscaled data. This evaluation was conducted using three disparate data sets to confirm the broad applicability of the approach, including: urine samples to discriminate between two dog breeds, muscle tissue extracts to discriminate between hypoxia and normoxia in marine mussels, and liver tissue extracts to discriminate between fish collected from two different rivers. The performances of each of the scaling methods – autoscaling, Pareto and glog – were assessed by conducting PCA of each of the processed twoclass data sets. This was achieved by calculating the sensitivities and specificities derived from applying linear discriminant analysis (LDA) to each of the resulting PCA scores plots. The effect of each scaling method upon the ability to discover potential metabolic biomarkers was also investigated. This was accomplished by selecting the largest peaks in the PCA loadings plots and then evaluating if the corresponding peaks in the NMR spectra were of significantly different intensity between the biological classes. Secondly, we aimed to evaluate the applicability of the glog transform for 1D NMR spectra, 1D projections of 2D JRES spectra, and intact 2D JRES spectra. This enabled the first NMR metabolomics study of intact 2D JRES spectra; including the reconstruction of the PCA loadings plot to a 2D format analogous to the JRES spectra, which is anticipated to have significant benefit in terms of the ease of metabolite identification. During this second aim we also sought to extend the glog transform to reduce the deleterious effects of noise.
Results and Discussion
Parameter values for all glog transformations.
Data type  Transform  λ value  y_{0} value 

1D NMR, mussel muscle  glog  2.0025 × 10^{8}   
extended glog  1.2689 × 10^{8}  8.7026 × 10^{5}  
pJRES NMR, dog urine  glog  2.3024 × 10^{9}   
extended glog  1.5175 × 10^{9}  4.9506 × 10^{5}  
2D JRES NMR, fish liver  glog  6.9974 × 10^{14}   
extended glog  4.0877 × 10^{12}  1.575 × 10^{6} 
Data Structure
Figure 1B shows the 1D skyline projection of a 2D JRES spectrum (termed pJRES), which is similar in appearance to the 1D spectrum. There is a large difference between the intensities of the smallest and largest peaks, again implying that a series of these spectra would show non constant variance. Finally, Figure 1C shows the contour plot of a typical spectrum from a 2D JRES NMR experiment. Here the peaks are less crowded than in the 1D spectra as they have been dispersed along a second dimension, and are symmetrically located about the 0 Hz line.
Prior to assessing the effects of scaling on variance, it is important to contrast the technical versus biological variability in the datasets. This can be achieved by calculating the median and range of the coefficients of variation (CV) for all the bins across a series of NMR spectra. Technical variability is measured by the CV of the technical replicates, and biological variability (which also includes technical variability) by the CV of the biological dataset. For the mussel 1D NMR data, the median CV of the technical replicates is 6.5% (range of 0.4–30.6%). In contrast, the median CV of the mussel biological data is 22.6% (range of 7.2–128.4%). Clearly the technical variance is a significant proportion of the biological variance, and therefore must be treated appropriately prior to multivariate analysis. Similar results are found for the two other data sets: the dog pJRES NMR data has median CVs of 13.4% (technical) and 52.1% (biological) with ranges of 0.6–70.4% and 14.6–272.1%, respectively. And for the fish 2D JRES NMR data the median CVs are 23.0% (technical) and 48.4% (biological) with ranges of 1.5–88.2% and 13.6–228.5%, respectively.
Effects of Scaling on Variance
Scaling methods can radically change the variance structure of the data set. Figure 3B shows the variance versus ranked mean for the mussel technical replicates after the spectra have been autoscaled. Here, the variance is now totally uniform, with the variance of every bin set to unity. This removes any bias that may arise due to the peak intensity. However, every bin is now treated equally, giving no differentiation between wanted signals representing peaks and unwanted signals such as noise. For Pareto scaled data (Figure 3C), there is considerable similarity in appearance to the unscaled data since there are a few bins with large mean intensity that clearly have a large variance. This structure is also repeated amongst the bins of lower mean intensity in a similar manner to the unscaled data (see insert, Figure 3C). This highlights that the Pareto scaled spectra also have a large range of variance throughout the data set, which would affect which bins contribute to the loadings in a PCA. Finally, Figure 3D shows the impact of the glog transform upon the variance of the spectra. Here the variance structure is very different from the other scaling methods as there is a wide range of bin intensities giving rise to bins with similar variances, demonstrating the variance stabilising effect of this transform. Although there are still bins with higher variances compared with the majority of bins, these bins are spread throughout the range of bin intensities and so removes any bias of PCA towards the largest peaks.
Effects of Scaling on Classification Accuracy
Classification statistics for each PCA model constructed.
Data type  Scaling  Sensitivity  Specificity  Correctly classified  Crossvalidation accuracy 

1D NMR, mussel muscle  unscaled  0.333  0.800  16 of 27  37.04% 
autoscaled  0.083  0.933  15 of 27  33.33%  
Pareto  0.500  0.733  17 of 27  51.85%  
glog  1.000  1.000  27 of 27  100.00%  
extended glog  1.000  0.86667  25 of 27  92.60%  
pJRES NMR, dog urine  unscaled  0.294  0.750  20 of 37  32.43% 
autoscaled  0.824  0.850  31 of 37  83.78%  
Pareto  0.530  0.700  23 of 37  56.76%  
glog  0.824  0.850  31 of 37  83.78%  
extended glog  0.824  0.850  31 of 37  83.78%  
2D JRES NMR, fish liver  unscaled  1.000  0.550  29 of 38  68.42% 
autoscaled  0.944  0.800  33 of 38  63.16%  
Pareto  0.944  0.800  33 of 38  86.84%  
glog  0.889  0.850  33 of 38  86.84%  
extended glog  1.000  1.000  38 of 38  100.00% 
Mussel adductor muscle samples
Figure 4 shows the PCA scores plots for the models generated from the 1D NMR spectra of mussel adductor muscle. The two classes correspond to muscle obtained from normally respiring animals and from hypoxic mussels. Clearly, the unscaled data has little classrelated structure, with samples of both classes intermingled with each other (Figure 4A). The decision boundary created by the LDA provides a benchmark of 16 of 27 samples correctly classified to compare against the scaled data sets (Table 2). For the autoscaled data set there is one predominant cluster of mixed samples and a single outlier (Figure 4B). Qualitatively, there is no improvement to the scores plot over the unscaled data, and the LDA decision regions show a slight decrease in accuracy, only correctly classifying 15 of the 27 samples. The Pareto scaled data appears similar to the unscaled samples, since there is no obvious discrimination between the two classes; 17 samples are correctly classified (Figure 4C). The scores plot for the glog transformed data set shows a totally different structure with complete separation of the two classes along PC2 (Figure 4D). The LDA forms a decision region which separates the two classes entirely, such that all 27 samples are correctly classified.
The 5 largest bins in each loadings plot have been tested as potential biomarkers using oneway ANOVAs. Clearly, as shown in Figures 5A, 5B and 5C for the unscaled, autoscaled and Pareto scaled data respectively, none of these bins are significantly different between the two classes and so are poor biomarkers. In contrast, only the glog transformed spectra yielded bins with the largest loadings that are all significantly different between the hypoxic and control animals (Figure 5D). This highlights a significant benefit of the glog transform for discovering useful and significant biomarkers from NMR metabolomics data.
Canine urine samples
For the pJRES NMR data set of urine samples from two breeds of dog, the processing methods show a similar effect upon the data (Figure 6). The PCA scores plot for the unscaled data shows little noticeable structure between the different classes, with the LDA classifier correctly identifying only 20 of the 37 samples (Figure 6A and Table 2). Pareto scaling performs only slightly better, also with no noticeable separation of the two classes, and only 23 samples correctly classified (Figure 6C). However, both the autoscaled (Figure 6B) and glog transformed (Figure 6D) data yield improved classifications of 31 of the 37 samples, with the same six samples being misclassified. The margin of separation between the two distinct clusters remains approximately the same for the autoscaled and glog transform analyses, giving no clear 'best' scaling method for this data set. The explanation behind the misclassification of the 6 samples is beyond the scope of this study, although it is important to realise that this potentially interesting result was only revealed when the data set was appropriately scaled to reduce the effects of technical variance.
Fish liver samples
An algorithm to increase the relative signal to noise ratio of the data was then investigated by extending the glog transformation to include an additional parameter, as shown in equation (4). Figure 9B shows the same 2D JRES spectrum after application of the extended glog transformation. The PCA scores plot of the extended glog transformed data is also changed (Figure 10A) compared to the original glog transformed data (Figure 8D), with the variance expressed by the first two PCs almost doubled to 12.1% and 6.9%. The most noteworthy result is that the reduction of noise using the extended glog transform also improves the LDA classifier, with all 37 of 37 samples now correctly assigned to their correct classes and separation between the two classes in PCA space now readily apparent (Figure 10A).
The corresponding PC1 loadings plot for the scores plot in Figure 10A is shown in two different orientations in Figures 10B (top view) and 10C (side view). When used in combination with the extended glog transformation the resulting loadings plot provides a powerful visualisation tool from which the metabolic differences between the two sample classes can be identified. In particular, the significant advantages over the more traditional 1D loadings plots derived from both 1D NMR data as well as 1D pJRES data [15] include the decreased congestion of peaks and the preservation of Jcoupling information. Ultimately this approach could increase the confidence of metabolite identification in NMR metabolomics.
Conclusion
We have demonstrated that autoscaling, Pareto scaling and the glog and extended glog transformations can significantly alter the variance structure of NMR metabolomics data, which in turn can improve the classification accuracy of multivariate models generated from the scaled data. This can help to extract important information from data sets, since improving the discrimination between sample classes can help to identify metabolic biomarkers. Specifically, we have demonstrated that the glog and extended glog transformations achieve the best, or equal best, classification accuracy compared to unscaled, autoscaled and Pareto scaled data on three example data sets. A classification accuracy of 100% was achieved for two data sets – the effect of hypoxia in invertebrate muscle extracts and the effect of sampling location on fish liver extracts – and an accuracy of 31 of 37 correctly classified for a third dataset examining breed discrimination using dog urine. Furthermore, from an analysis of the top five peaks in each of the corresponding PCA loadings plots, we have confirmed that glog transformed data is considerably better at discovering metabolic biomarkers that can discriminate significantly between sample classes. We have also confirmed the broad applicability of the glog approach using three disparate data sets from different biological samples using 1D NMR spectra, 1D projections of 2D JRES spectra, and intact 2D JRES spectra. Finally, we have reported an extension to the original glog algorithm that effectively suppresses the noise, which was critical for the analysis of intact 2D JRES spectra. In conclusion, we have thoroughly evaluated and proven the benefits of utilising the glog transformation for stabilising the technical variance associated with metabolomics experiments, which can lead to significantly beneficial effects on the discrimination between sample classes using multivariate analysis.
Methods
Three data sets were used to highlight the broad applicability of the generalised log transformation across multiple biological species and sample types. The three data sets comprised spectra of mammalian (canine) urine, extracts of marine mussel adductor muscle, and extracts of fish liver. The preparation, NMR analysis and processing of each is described below.
Sample Preparation and Collection of NMR Spectra
Canine urine
Freecatch urine samples were collected over several days from two breeds of dog (17 samples from three male Labradors and 20 samples from four male Miniature Schnauzers), frozen at 80°C, and subsequently prepared and analysed using the methods described elsewhere [22]. Briefly, urine was diluted in a sodium phosphate buffer (pH 7.0; 100 mM final concentration) containing sodium 3trimethylsilyl2,2,3,3d4propionate (TMSP; 1 mM final concentration), 0.2% sodium azide and 8% D_{2}O. The sample pH was then manually adjusted to 7.05 (± 0.05) using 1 M HCl or NaOH. Samples were analyzed on a Bruker 500 MHz NMR spectrometer equipped with a 5 mm cryoprobe and BACS60 automatic sample changer. 2D ^{1}H, ^{1}H JRES NMR spectra were collected with 16 increments using excitation sculpting to suppress the water resonance, and transients were processed using methods described previously [15], yielding 1D skyline projections of the JRES spectra (termed pJRES). In addition to the 37 individual urine samples, an additional pooled sample was split into 5 fractions and each of these was then prepared and analysed separately, using the same methods as above. This provided the spectra of technical replicates needed to calibrate the generalised log transform.
Mussel adductor muscle
Muscle tissues were dissected from two groups of Mediterranean mussels (Mytilus galloprovincialis), the first group being hypoxic (i.e. oxygen deficient; n = 12) and the second group normoxic (n = 15). The tissues were prepared using a methanol:chloroform extraction protocol as recently reported [23]. Polar extracts were dried and then resuspended in 100 mM sodium phosphate buffer (pH 7.0; 1 mM TMSP; 10% D_{2}O). 1D ^{1}H NMR spectra of the polar metabolites were collected, as described previously [24]. Similar to the canine urine study, an additional pooled tissue sample was homogenised, split into 6 fractions and then each fraction was extracted and analysed separately, providing spectra of technical replicates.
Fish liver
European flounder (Platichthys flesus) were sampled from the River Alde, UK (n = 20) and the River Tyne, UK (n = 18). Liver tissue was rapidly dissected and then extracted using the methanol:chloroform protocol as above [23]. All polar extracts were dried and resuspended in 90% H_{2}O and 10% D_{2}O with sodium phosphate buffer (100 mM; pH 7.0) containing 0.5 mM TMSP. 2D ^{1}H, ^{1}H JRES NMR spectra were collected using methods described above. Again, an additional pooled tissue sample was homogenised, split into 5 fractions and then each was extracted and analysed separately, providing spectra of technical replicates.
Technical Replicates
It should be noted that for all data sets, the technical replicates form an integral part of calibrating the glog transformation. A minimum of five or six replicates should be generated from a single homogenous pool of the relevant biological material for each data set. Ideally, this pool of biological material is formed by mixing several smaller amounts of different samples from all experimental classes (e.g., control and stressed).
Data Processing
The 1D, pJRES and 2D JRES NMR spectra were converted to an appropriate format for multivariate analysis using customwritten ProMetab software [15] running within MATLAB (version 7.1; The MathWorks, Natick, MA). All spectra were sectioned into 1960 chemical shift bins between 0.2 and 10.0 ppm, corresponding to a bin width of 0.005 ppm. Note that the 2D JRES spectra were not "binned" along the J coupling dimension at this stage of the processing. Next, a series of bins were removed from each data set: for the canine urine from 4.50–6.45 ppm (residual water and urea); for the mussel adductor muscle from 4.70–5.15 ppm (residual water) and 7.60–7.76 ppm (chloroform); and for fish liver from 4.60–5.20 ppm (residual water). The spectra for each data set were then normalised to a total spectral area of unity for ease of comparison between samples. Next, due to slight pHinduced chemical shift variations of some peaks between samples, groups of bins were each compressed into single bins: for the canine urine ten regions were compressed between 2.40–2.425, 2.52–2.57, 2.66–2.71, 2.935–2.955, 2.96–2.98, 3.105–3.130, 3.72–3.77, 3.955–3.990, 7.08–7.20 and 8.00–8.18 ppm; for the mussel adductor muscle between 7.08–7.10 and 7.84–7.875 ppm; and for fish liver five regions were compressed between 7.74–7.77, 7.77–7.79, 7.94–7.955, 7.97–8.03 and 8.23–8.25 ppm. Compression regions were chosen by visually inspecting the superimposed NMR spectra and then selecting regions of the spectra that showed pH or matrix induced chemical shift variation. Finally, for the fish liver only, the increments of each intact 2D JRES spectrum (i.e. the rows of the 2D data matrix representing each spectrum) were concatenated into a single row vector of dimension 232,448 containing the intensities of each bin in the spectrum, allowing the JRES spectra to be analysed in a similar manner to the 1D and pJRES spectra, described below.
Scaling Methods
After each data set was binned, normalised and bin compressed – and for the intact 2D JRES spectra, concatenated – the following scaling techniques were applied:
Autoscaling
The variance of each bin was scaled to unity by dividing the intensity of each bin by the standard deviation of that bin; note that mean centring was not applied yet.
Pareto scaling
The intensity of each bin was divided by the square root of the standard deviation of that bin; again, mean centring was not applied at this point.
Glog transformation
Here, $\widehat{w}$ is calculated as the mean spectrum of all scaled and transformed technical replicates, w_{ j }. Minimising the variance S thus gives an optimal value for λ.
The optimisation of λ is achieved via the NelderMead unconstrained nonlinear minimization routine in the MATLAB optimisation toolbox. The optimised λ value was then used to transform the binned intensities of each spectrum in the full biological data set. The MATLAB code developed here is included as additional file 2.
Since y_{0} depends on the choice of λ the optimisation of λ must be carried out first followed by the calculation of y_{0}. In some cases it may be necessary to optimise λ a second time after y_{0} has been set, in particular for very noisy data.
For both calibration methods described here, the minimisation routine was terminated when the absolute change in λ was less than a predetermined value (here 1 × 10^{16}) or a maximum number of iterations was completed (here 1 × 10^{3}). Table 1 contains the optimised λ and y_{0} values for the glog and extended glog transformations for each of the three biological samples investigated.
Analysis of Models
Each unscaled or scaled data set was then mean centred and PCA performed using PLS_Toolbox (Eigenvector Research, Inc., Wenatchee, WA, USA). Next, using the Discriminant Analysis Toolbox (Michael Kiefte, Dalhousie University, Canada [27]) Fisher's LDA was applied to the first and second PCs of the PCA scores plot, producing a decision region for each twoclass problem. This decision region was then used to construct classification statistics (sensitivities and specificities) to evaluate the effects of the scaling techniques upon each data set (Table 2). Leaveoneout crossvalidation was performed on the PCALDA models to assess the robustness of the analyses (Table 2). The coefficient of variation (CV) for each bin, given as the standard deviation divided by the mean, was calculated for each set of technical replicates, excluding bins with an intensity lower than the estimated noise level of the spectrum (i.e., the CV was calculated using only those bins that contained peaks). The median and range of these CVs were calculated for each of the three data sets. Additionally, PCA loadings plots for the 1D and pJRES data (Figures 5 and 7) were produced by constructing the linear combination of the loadings along PC1 and PC2 that is perpendicular to the LDA decision line. The loadings plot for the 2D JRES experiment, shown in 2D matrix format to mimic an intact 2D JRES spectrum (Figures 10B and 10C), was reconstructed from the row vector containing the loadings of the concatenated spectra. To evaluate the discriminatory potential of metabolic biomarkers discovered in the loadings plots, oneway analysis of variance (ANOVA) was performed on each of the 5 bins with the largest absolute loadings values, for each data set and method of scaling.
Notes
List of abbreviations used
 NMR:

nuclear magnetic resonance
 PCA:

principal component analysis
 PLSDA:

partial least squares discriminant analysis
 PC:

principal component
 LDA:

linear discriminant analysis
 glog:

generalised logarithm transformation
 1D:

one dimensional
 2D:

two dimensional
 JRES spectrum:

2D Jresolved NMR spectrum
 pJRES:

1D skyline projection of a 2D JRES spectrum
 ANOVA:

analysis of variance
 CV:

coefficient of variance
Declarations
Acknowledgements
HMP thanks the EPSRC and NERC for a Directed PhD studentship and MRV thanks the NERC for an Advanced Fellowship (NER/J/S/2002/00618). This work was partly supported by the NERC Post Genomic and Proteomic (PGP) Directed Program (NE/C507661/1). The authors gratefully acknowledge Prof. David Rocke, Prof. David Woodruff, Yuanxin Xi (University of California, Davis) and John Easton (Birmingham) for assistance with the MATLAB code, as well as Dr. Dov Stekel (Birmingham) for advice on the linear discriminant analyses. We also thank several people for supplying samples and/or NMR data, including Dr. David Allaway (Waltham Centre for Pet Nutrition) for the canine urine samples, Dr. Stephen George (University of Stirling) for the flounder liver samples, and Dr. Huifeng Wu and Adam Hines (Birmingham) for many of the NMR spectra.
Authors’ Affiliations
References
 Goodacre R, Vaidyanathan S, Dunn WB, Harrigan GG, Kell DB: Metabolomics by numbers: acquiring and understanding global metabolite data. Trends Biotechnol 2004, 22: 245–252. 10.1016/j.tibtech.2004.03.007View ArticlePubMed
 van den Berg Robert HH, Johan W, Age S: Centering, scaling, and transformations: improving the biological information content of metabolomics data. BMC Genomics 2006, 7: 142. 10.1186/147121647142PubMed CentralView ArticlePubMed
 Eriksson L, Johansson E, KettanehWold N, Wold S: Multiand megavariate data analysis: principles and applications. Umetrics; 2001.
 Beckonert O, Bollard ME, Ebbels TMD, Keun HC, Antti H, Holmes E, Lindon JC, Nicholson JK: NMRbased metabonomic toxicity classification: hierarchical cluster analysis and knearestneighbour approaches. Anal Chim Acta 2003, 490: 3–15. 10.1016/S00032670(03)000606View Article
 Lindon JC, Holmes E, Nicholson JK: Pattern recognition methods and applications in biomedical magnetic resonance. Progress in Nuclear Magnetic Resonance Spectroscopy 2001, 39: 1–40. 10.1016/S00796565(00)000364View Article
 Jones GLAH, Sang E, Goddard C, MortishireSmith RJ, Sweatman BC, Haselden JN, Davies K, Grace AA, Clarke K, Griffin JL: A Functional Analysis of Mouse Models of Cardiac Disease through Metabolic Profiling. Journal of Biological Chemistry 2004, 280: 7530–7539. 10.1074/jbc.M410200200View ArticlePubMed
 Ripley BD: Pattern Recognition and Neural Networks. Cambridge University Press; 1996.View Article
 Churchill GA: Fundamentals of experimental design for cDNA microarrays. Nature Genetics 2002, 32: 490–495. 10.1038/ng1031View ArticlePubMed
 Craig A, Cloarec O, Holmes E, Nicholson JK, Lindon JC: Scaling and Normalization Effects in NMR Spectroscopic Metabonomic Data Sets. Analytical chemistry(Washington, DC) 2006, 78(7):2262–2267.
 Keun HC, Ebbels TMD, Antti H, Bollard ME, Beckonert O, Holmes E, Lindon JC, Nicholson JK: Improved analysis of multivariate data by variable stability scaling: application to NMRbased metabolic profiling. Analytica Chimica Acta 2003, 490: 265–276. 10.1016/S00032670(03)000941View Article
 Purohit PV, Rocke DM, Viant MR, Woodruff DL: Discrimination models using variancestabilizing transformation of metabolomic NMR data. OMICS 2004, 8: 118–130. 10.1089/1536231041388348View ArticlePubMed
 Rocke DM, Durbin B: Approximate variancestabilizing transformations for geneexpression microarray data. Bioinformatics 2003, 19: 966–972. 10.1093/bioinformatics/btg107View ArticlePubMed
 Geller SC, Gregg JP, Hagerman P, Rocke DM: Transformation and normalization of oligonucleotide microarray data. Bioinformatics 2003, 19: 1817–1823. 10.1093/bioinformatics/btg245View ArticlePubMed
 Durbin BP, Hardin JS, Hawkins DM, Rocke DM: A variancestabilizing transformation for geneexpression microarray data. Bioinformatics 2002, 18: S105S110.View ArticlePubMed
 Viant MR: Improved methods for the acquisition and interpretation of NMR metabolomic data. Biochemical and Biophysical Research Communications 2003, 310: 943–948. 10.1016/j.bbrc.2003.09.092View ArticlePubMed
 Forveille L, Vercauteren J, Rutledge DN: Multivariate statistical analysis of twodimensional NMR data to differentiate grapevine cultivars and clones. Food Chemistry 1996, 57: 441–450. 10.1016/03088146(95)002200View Article
 Dumas ME, Canlet C, André F, Vercauteren J, Paris A: Metabonomic assessment of physiological disruptions using 1H13C HMBCNMR spectroscopy combined with pattern recognition procedures performed on filtered variables. Anal Chem 2002, 74: 2261–2273. 10.1021/ac0156870View ArticlePubMed
 Widarto HT, Van Der Meijden E, Lefeber AWM, Erkelens C, Kim HK, Choi YH, Verpoorte R: Metabolomic Differentiation of Brassica rapa Following Herbivory by Different Insect Instars using TwoDimensional Nuclear Magnetic Resonance Spectroscopy. Journal of Chemical Ecology 2006, 32: 2417–2428. 10.1007/s1088600691526View ArticlePubMed
 Liang YS, Choi YH, Kim HK, Linthorst HJ, Verpoorte R: Metabolomic analysis of methyl jasmonate treated Brassica rapa leaves by 2dimensional NMR spectroscopy. Phytochemistry 2006, 67: 2503–2511. 10.1016/j.phytochem.2006.08.018View ArticlePubMed
 Wang Y, Bollard ME, Keun H, Antti H, Beckonert O, Ebbels TM, Lindon JC, Holmes E, Tang H, Nicholson JK: Spectral editing and pattern recognition methods applied to highresolution magicangle spinning 1H nuclear magnetic resonance spectroscopy of liver tissues. Anal Biochem 2003, 323: 26–32. 10.1016/j.ab.2003.07.026View ArticlePubMed
 Viant MR, Bundy JG, Pincetich CA, de Ropp JS, Tjeerdema RS: NMRderived developmental metabolic trajectories: an approach for visualizing the toxic actions of trichloroethylene during embryogenesis. Metabolomics 2005, 1: 149–158. 10.1007/s1130600544292View Article
 Viant MR, Ludwig C, Rhodes S, Günther UL, Allaway D: Validation of a urine metabolome fingerprint in dog for phenotypic classification.
 Lin CY, Wu H, Tjeerdema RS, Viant MR: Evaluation of Metabolite Extraction Strategies From Tissue Samples Using NMR Metabolomics. Metabolomics 2007, 3: 55–67. 10.1007/s1130600600431View Article
 Hines A, Oladiran GS, Bignell JP, Stentiford GD, Viant MR: Direct Sampling of Organisms from the Field and Knowledge of their Phenotype: Key Recommendations for Environmental Metabolomics. Environmental Science & Technology 2007, 41: 3375–3381. 10.1021/es062745wView Article
 Durbin B, Rocke DM: Estimation of transformation parameters for microarray data. Bioinformatics 2003, 19: 1360–1367. 10.1093/bioinformatics/btg178View ArticlePubMed
 Golotvin S, Williams A: Improved Baseline Recognition and Modeling of FT NMR Spectra. J Magn Reson 2000, 146: 122–125. 10.1006/jmre.2000.2121View ArticlePubMed
 Kiefte M: Discriminant Analysis Toolbox.[http://www.mathworks.com/matlabcentral/fileexchange/loadFile.do?objectId=189&objectType=FILE]
Copyright
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.