Highly sensitive feature detection for high resolution LC/MS
© Tautenhahn et al; licensee BioMed Central Ltd. 2008
Received: 09 July 2008
Accepted: 28 November 2008
Published: 28 November 2008
Liquid chromatography coupled to mass spectrometry (LC/MS) is an important analytical technology for e.g. metabolomics experiments. Determining the boundaries, centres and intensities of the two-dimensional signals in the LC/MS raw data is called feature detection. For the subsequent analysis of complex samples such as plant extracts, which may contain hundreds of compounds, corresponding to thousands of features – a reliable feature detection is mandatory.
We developed a new feature detection algorithm centWave for high-resolution LC/MS data sets, which collects regions of interest (partial mass traces) in the raw-data, and applies continuous wavelet transformation and optionally Gauss-fitting in the chromatographic domain. We evaluated our feature detection algorithm on dilution series and mixtures of seed and leaf extracts, and estimated recall, precision and F-score of seed and leaf specific features in two experiments of different complexity.
The new feature detection algorithm meets the requirements of current metabolomics experiments. centWave can detect close-by and partially overlapping features and has the highest overall recall and precision values compared to the other algorithms, matchedFilter (the original algorithm of XCMS) and the centroidPicker from MZmine. The centWave algorithm was integrated into the Bioconductor R-package XCMS and is available from http://www.bioconductor.org/
Metabolomics aims at the unbiased and comprehensive quantification of metabolite concentrations in organisms, tissues, or cells [1, 2]. The combination of chromatographic separation with subsequent mass spectrometric detection has emerged as a key technology for multiparallel analysis of low molecular weight compounds in biological systems. Gas chromatography-mass spectrometry (GC/MS) based techniques are mature and well-established, but restricted to volatile compounds, often requiring chemical derivatisation. High-performance liquid chromatography-mass spectrometry (HPLC/MS) facilitates the analysis of compounds of higher polarity and lower volatility in a much wider mass range without derivatisation [3–5]. With LC/MS the injected sample is separated on the chromatographic column, resulting in the consecutive elution of different compounds. The mass spectrometer acquires mass spectra from the column output at a specified scan rate, so each compound can be measured in several consecutive scans. Due to the fact that each eluting compound gives rise to a number of mass signals (adducts, fragments and isotopic peaks), a metabolite induces several two-dimensional features.
In the following, we use the term "feature" for a bounded, two-dimensional (m/z and retention time) LC/MS signal. The term "peak" is used for one-dimensional signals: both m/z peaks (centroids) in the mass spectrum and chromatographic peaks.
For complex metabolomics samples, the LC/MS data contains hundreds to thousands of metabolites. For the statistical analysis of biological experiments the feature intensity is of interest and has to be calculated from the raw data. Spectra can be acquired in profile mode or centroid mode. Vendor supplied centroidisation algorithms usually employs machine-specific models, which are superior to generic approaches. In addition, the centroid mode results in considerable size reduction of the LC/MS data set.
Signal preprocessing and centroidization in m/z,
Two-dimensional feature detection and integration
Alignment of corresponding features in multiple samples
Statistical analysis, chemical and biological interpretation.
Feature detection is a crucial step in the LC/MS data processing pipeline – it should be reliable, i.e. report as many as possible "real" features, while keeping the false positive rate low. The challenge for the algorithms is to detect features of low intensity induced by compounds with low abundance on the one hand, and to avoid feature-like signals caused by e.g. chemical noise on the other hand.
Several frameworks for feature detection (and alignment) of metabolomics LC/MS data have been developed in the last years, both commercial products such as MarkerLynx (Waters), the closed-source (but freely-available) MetAlign , or XCMS  and MZmine  which have open-source licenses. Other packages, some of them specific for LC/MS-based proteomics, have been reviewed in .
A widely used approach for the processing of LC/MS data is to transform the raw data into a matrix representation with the dimensions m/z, retention time and intensity. To convert high resolution mass spectra into this representation, it is necessary to divide the m/z axis into equidistant chunks depending on the resolution of the mass spectrometer, e. g. 0.1 m/z wide. This procedure is usually referred to as binning. Some drawbacks of this method were already mentioned in [7, 10, 11]. In particular, specifying the optimal bin size for the particular data set can be difficult. If the bin size is chosen too small, chromatographic peaks are alternating between bins and cannot be detected due to the loss of the chromatographic shape. If the bin size is too large, peaks can overlay each other and small features are rather buried by the increased chromatographic noise level. On the positive side it should be mentioned that the binning approach is all-purpose and allows for a fast data processing.
A density based LC/MS feature detection approach – an alternative to the common binning technique – was introduced by Stolt et al. . The authors consider the emerging analyte as a region of data points with high density anked by a specific "data void". Based on these properties, they calculate a potential field which is then used to create a matrix of mass traces (runtime ~2 h/sample). Recently, the extraction of "pure ion chromatograms" using Kalman tracking was demonstrated in . The applicability of Wavelet based techniques for peak picking in MALDI- and SELDI-TOF mass spectra was shown by e.g. [12–15]. Here we will discuss a new method for the reliable detection and integration of two-dimensional LC/MS signals, referred to as features. By using a combination of a density based technique to detect regions of interest in the m/z domain, and a Wavelet based approach to resolve chromatographic peaks, we achieve a high sensitivity even in very complex mixtures compared to two other algorithms, matchedFilter (the original algorithm of XCMS) and the centroidPicker from MZmine.
So far, there is no common method for evaluating the performance of feature detection algorithms. Even for the same feature detection algorithm, different parametrisation can lead to (vastly) different results, if e.g. many false positive noise signals are detected as features. Therefore the absolute number of detected features per sample is not suitable to characterise a feature detection algorithm. More elaborate approaches consider mixtures of known compounds spiked into complex samples . To the best of our knowledge, no evaluation has been performed to assess recall and precision of feature detection algorithms for multiple complex samples.
The remainder of this paper is structured as follows: In section 2 we give a detailed description of the centWave algorithm, followed by the description of the experimental comparison between several feature detection algorithms. In section 3 we present the evaluation results and discuss the benefits of centWave, followed by a conclusion and outlook of expected future developments in section 4.
This section describes the centWave method which combines density based detection of regions of interest in the m/z domain, and a Continuous Wavelet Transform (CWT) based approach for chromatographic peak resolution. The experimental setup is depicted as well as the layout of the evaluation procedure.
2.1 The centWave algorithm
2.1.1 Detecting regions of interest (ROI) in the m/z domain
Due to the fact that the mass accuracy (μ, given in ppm) of the mass spectrometer and the minimum chromatographic peak width is known or can easily be assessed, it is possible to directly scan for regions where at least pmin centroids with a deviation less than μ ppm occur. This task is achieved by the following algorithm for samples in centroid mode, with scans numbered s = 1,...,S:
(a) Initialise a list ROI using all m/z values from the first scan:
∀ i = 1,..., N, N = |mzs = 1|: ROI(i).values(1) =
(b) Initialise the m/z mean value for each actually processed region :
ROI(i).mzmean = , i = 1,..., N, N = |mzs = 1|
2. For each scan s = 2,..., S :
(a) For each m/z value , i = 1,..., N, N = |mz s | in the current scan s:
Exists j, j = 1,..., J, J = |ROI| such that |ROI(j).mzmean - | < = μ ?
• Yes: Append to ROI(j) and update the m/z mean value
K = |ROI(j).values| + 1, ROI(j).values(K) =
ROI(j).mzmean = .values(k)
• No: Initialise a new ROI and append it to the list
Check & Cleanup:
Remove all ROI which were not extended in step 2a and contain less than pmin centroids
Mark ROI that were not extended, but contain at least pmin centroids as completed
Optionally an intensity filter (prefilter = (k, I), e. g. prefilter = (2, 100)) can be set to early discard regions of small intensity. Then only those ROI are retained (in step 2b) that contain at least k consecutive values with intensity ≥ I. This prefilter vastly speeds up the overall processing time.
In some rare cases "gaps" are observed in the mass trace of features with low intensity. Due to the fact that each ROI is laterally extended for the following chromatographic peak detection, only a small contiguous region needs to be found for the successful detection of such features. To a certain extent, the algorithm is therefore able to detect features with such gaps. Otherwise, in case of samples which might show this phenomenon more often, the algorithm can easily be modified to be even more "gap-tolerant". In contrast to binning, this approach has the advantage that no fixed bin size has to be chosen. Each ROI is detected separately and the drawbacks of binning can be circumvented. Unlike binning the result is not a matrix but a list of mass traces with different lengths. Depending on the chromatography and the mass accuracy of the mass spectrometer, each ROI may contain none, exactly one or more than one distinct chromatographic peaks. Therefore it is necessary to subject each ROI to an extensive analysis in the chromatographic domain.
2.1.2 Detecting chromatographic peaks
2.1.3 The centWave workflow
Mass deviation μ in ppm, typically set to a generous multiple of the mass accuracy of the mass spectrometer. We use μ = 30 ppm for the Bruker MicrOTOF-Q, which is advertised with a mass accuracy of 3–5 ppm.
Chromatographic peak width range wmin, wmax in seconds, e. g. wmin, wmax = (5, 10) for UPLC separation as described in the experimental setup.
Signal to noise ratio threshold SNRThr, e.g. SNRThr = 10
The following is the description of the most important steps of the centWave workflow:
The scale range smin, smax for the CWT and the pmin parameter for the ROI detection are calculated from the input parameters wmin, wmax and the average inter-scan distance.
ROI detection (see section 2.1.1) is performed using the parameters μ and pmin
Chromatographic analysis of each detected ROI:
To accommodate noise and baseline estimation, each ROI is laterally extended by a multiple of the expected chromatographic peak width
Local noise and baseline estimation: Let x be the vector of intensity values of the actual (extended) ROI, and x t the 10% trimmed x (5% of the smallest and 5% of the largest intensity values are discarded). Then the baseline BL is assessed as the mean value of x t and the noise level NL as the standard devation of x t .
The Continuous Wavelet Transform (see 2.1.2) is applied to the intensity values of the ROI (the extracted ion chromatogram), using the scale range smin,..., smax.
Local maxima of the CWT coefficients at each scale are detected.
"Ridges" can be identified by linking the detected local maxima (described in ). The ridges describe the scale range where the chromatographic peak was located. If more than one chromatographic peak was detected, the following steps are applied for each peak separately.
Locate the chromatographic peak boundaries rtmin and rtmax by descent on the filtered peak data, i.e. the CWT coefficients of the scale where the peak was optimally located.
Calculate the feature intensity I using the intensity values within rtmin and rtmax. Imax is defined as the maximal intensity value within this range.
Compute the m/z centroid of the feature as the weighted mean of the m/z values within rtmin and rtmax.
Calculate the signal to noise ratio SNR = (Imax – BL)/NL of the feature. Discard the feature if SNR < SNRThr.
The deviation μ* of m/z values within rtmin and rtmax is calculated in ppm. The value μ* can be interpreted as the mass deviation value which would have been sufficient for the detection of this feature.
Optionally, a Gaussian curve is fitted to the feature, using the Nonlinear Least Squares (NLS) implementation of R.
2.2 Experimental setup and Sample description
Arabidopsis thaliana (ecotype Col-0) was grown under controlled conditions and pooled after harvest. Methanolic extracts were prepared from ground seed and leaf tissue. o-Anisic acid, biochanin A, p-coumaric acid, ferulic acid, N-(3-indolylacetyl)-L-valine, kinetin, indole-3-acetonitrile, indole-3-carbaldehyde, kaempferol, phloretin, phlorizin and phenylglycine, rutin, and phenylalanine-d5 were used as marker compounds. The chromatographic separations were performed on an Acquity UPLC system (Waters) equipped with a modified C18 column with a 20 min water/acetonitrile gradient. The eluted compounds were detected by a Bruker MicrOTOF-Q in positive ion mode at a scan rate of 3 Hz. Mass calibration was performed against lithium formiate. The detailed experimental setup is available as Additional file 1.
Sample 1 A mixture containing each of the fourteen marker compounds (referred to as MM14) at a concentration of 20 μ M was prepared and analysed by UPLC/ESI-QTOF-MS.
Sample set 2 Mixtures containing solvent and seed or leaf extracts were prepared with following volume portions (solvent/seed/leaf, v/v/v): 0/100/0, 25/75/0, 50/50/0, 75/25/0, 0/0/100, 25/0/75, 50/0/50, 75/0/25. The sample set (8 samples) was analysed by UPLC/ESI-QTOF-MS in ten technical replications.
Sample set 3 Mixtures containing solvent, seed, and leaf extracts were prepared with following volume portions (solvent/seed/leaf, v/v/v): 75/0/25, 0/75/25, 0/50/50. The sample set (3 samples) was analysed by UPLC/ESI-QTOF-MS in ten technical replications.
All files were acquired in centroid mode and converted to mzData file format using Bruker CompassXport software. The data sets are available at http://msbi.ipb-halle.de/msbi/centwave/.
2.3 Parameter optimisation
Overview of the evaluated feature detection algorithms
The three algorithms tested have a number of parameters each, which have to be tuned to deliver good performance on the analytical setup. The centWave algorithm uses the peakwidth (= wmin, wmax) parameter to specify the chromatographic peak width range, the ppm parameter to set the tolerated mass deviation and snthresh, which defines the chromatographic signal-to-noise threshold. The matchedFilter algorithm has a similar parameter snthresh, the chromatographic peak width is specified by the fwhm parameter, which defines the width of the model peak for matched filtering. The mass accuracy is indirectly defined by the bin size (parameter step).
The centroidPicker from MZmine also needs a bin size to be specified (bin size), and additionally the tolerated mass deviation (m/z tolerance). Moreover, there are five parameters that affect the chromatographic domain: chromatographic threshold level, intensity tolerance, minimum peak duration, minimum peak height and noise level. The first two of those are specified as a relative value, while the last three are set using absolute values.
The parameters of the three algorithms were tuned to detect as many of the real features, without allowing too many false positives. Based on known "good working" settings, we performed parameter sweeps and evaluated the number of real features and the number of other ("false") features for each setting. After initial optimisation of the other parameters, we found that for centWave and matchedFilter the snthresh parameter shows the highest influence on this ratio.
The centroidPicker from MZmine was more complex to optimise, due to its many parameters. Using settings from the authors as a starting point, a sweep was performed over a wide parameter range. Approximately 500 parameter settings were tried for MZmine, and about 50 for matchedFilter and centWave.
For the parameter optimisation we used the mixture of 14 compounds (MM14). Due to the electrospray ionisation, each compound gives rise to a number of features. A data set of known features was created using the separately measured substances. We annotated features that can be explained as adducts and fragments of the compound as well as their isotopic peaks. For all 14 compounds this results in a set of 296 features, about 21 features per compound. We observed up to eight in-source fragments per compound and also various cluster ions like [2M+H]+ oder [3M+Na]+. The annotations are available as Additional file 2. Manual verification shows, that 122 of the 296 known features are clearly visible in the MM14 mixture, while the other 174 features are hard to detect by the human eye. The 122 verified features are considered as required features, which should be detected by the algorithms.
All other features (beyond the 296) which were reported by the programs, but cannot be explained as features originating from the marker mixture, are considered as "false" features, e.g. (usually small) signals from solvents or chemical impurities, background noise etc.
As one result from the optimisation, we found that all algorithms are able to detect more than 100 from the 122 selected real features, but only if approximately 450 "false" features are tolerated. The total number of 122 real features are detected only with settings that give more false positives (see section 3.4). Therefore, as a trade-off between real and "false" features, we chose those parameter settings which detect a maximal number of real features, but return less than 450 "false" features.
Since the algorithms detect around 200–300 features in the separately measured blank solvent, these 450 "false" features can be explained as a "background", consisting of features originating from the solvents, tubes, vials, or impurities of the used marker compounds.
Parameter optimisation using the MM14 marker mixture
Number of detected MM14 features
Number of other reported features
peakwidth = (5,10), ppm = 30, snthresh = 5, prefilter = (2,400)
fwhm = 4, snthresh = 12, step = 0.02, mzdiff = 0, max = 50
bin size = 0.05, chromatographic threshold level = 0.8, intensity tolerance = 0.7, minimum peak duration = 3, minimum peak height = 500, m/z tolerance = 0.03, noise level = 20
Since feature detection can be seen as an information retrieval task, the performance can be assessed using the precision and recall values. The recall value (also referred to as sensitivity) measures the fraction of relevant items that are found by a query, while the precision value quantifies the relation of relevant items to the false positives. Denoting the total number of features that were detected by an algorithm by N, the number of real features that were found by TP, and the total number of real features by NP, we can measure Recall = and Precision = of a feature detection algorithm. A perfect feature detection algorithm will have both measures equal to 100%. False positives features lower the precision; false negatives (undetected real features) lower the recall.
For a compact representation of the results we used the F-score as a combined measure of precision and recall, which is defined as F-score = . A perfect feature detection will achieve a F-score of 100%, and both false positives and false negatives features lower its value. The F-score can be interpreted as a measure of the overall performance of a feature detection algorithm.
Results and discussion
We performed two experiments to assess the performance of the three algorithms. The experiments were designed to evaluate the sensitivity of the algorithms using complex biological samples at different concentrations.
First, the feature set representing the ground truth had to be created. For this purpose we used ten technical replicates of undiluted Arabidopsis thaliana seed and leaf extracts from Sample set 2 (solvent/seed/leaf): (0/100/0) and (0/0/100).
Feature detection on the 2 × 10 samples was performed using the three algorithms
We investigated the number of features which are found reproducibly in repeated measurements. The features detected in the ten technical replicates of undiluted seed and leaf extracts were separately aligned using XCMS group function (mzwid = 0.05, bw = 2). After the alignment only those features which were detected in at least seven out of the ten samples were retained. The resulting numbers of features are shown in Table 3.
We matched the aligned feature lists of all three algorithms (using 0.015 m/z and 5 s tolerance) and removed those features which had been found by only a single algorithm.
Number of aligned features
3.1 Experiment 1
3.2 Experiment 2
For the second experiment we created mixtures of the seed and leaf extract at different concentrations (Sample set 3) and evaluated the F-score of the ground truth features. Again, feature detection was performed with the three algorithms. The ground truth seed and leaf specific features were considered together as true positives for this measurement. Thereby, the features which appear in both, seed and leaf extracts, were considered only once. All other features that were returned by the algorithms were considered as false positives. The right-most part of Figure 9 shows the result.
The detailed F-score, recall, and precision values of both experiments are available as Additional file 3. By manual inspection of the "true" features that were detected by centWave, but not by MZmine or matchedFilter, we found that these features were often close to other – in many cases larger – chromatographic peaks. This can be interpreted as a masking effect caused by noise level computation on the full chromatogram. The centWave algorithm uses local baseline and noise estimation to circumvent this problem.
Looking at the false positive features, we observed that matchedFilter frequently reports spikes (very narrow chromatographic peaks, consisting of 1–3 points) while MZmine tends to detect features in regions where only a high level noise can be seen.
Runtime in minutes
3.4 Alternative parameter settings
The optimisation strategy outlined in section 2.3 tried to balance the recall and precision, using a rough estimate of the potential chemical background signals. The settings in typical metabolomics profiling experiments of e.g. plant extracts usually will be tuned more aggressively towards higher sensitivity. The resulting false positive features are often filtered by the downstream analysis, such as the alignment of replicate measurements and statistical tests for differential features.
With the second parameter set, we observed higher sensitivity for all algorithms. The number of aligned features almost doubled, the resulting ground truth contains 6649 unique features. The recall values of matchedFilter improved notably with the alternative parameter settings.
The results based on the second parameter set confirm the general trend shown above. The centWave algorithm achieved up to 6% and 15% higher F-score values than MZmine and matchedFilter, respectively.
We presented a new feature detection algorithm for high resolution LC/MS data called centWave. With the increasing deployment of high-resolution mass spectrometers such as QTOF or Orbitrap instruments, and high-throughput applications such as metabolomics experiments of highly complex samples, a reliable and sensitive feature detection is essential. centWave shows a high sensitivity, while trying to keep the false positive features low.
In the past, the Bioconductor project has attracted more and more development related to mass spectrometry and metabolite pathways. The implementation of centWave is available in the R-package XCMS and can be obtained from http://bioconductor.org/packages/release/bioc/html/xcms.html. Integration with Bioconductor provides good support for the common file formats (netCDF, mzData and mzXML, with mzML currently under development) and allows for powerful downstream statistical analysis. The user feedback on the XCMS mailing list showed, that the centWave algorithm (introduced in 2007) is successfully used for LC-QTOF, LC-Orbitrap and even CE-MS or GC-MS data. For a more objective comparison we have evaluated centWave against two other open source algorithms. We performed two experiments to assess the performance of the algorithms, using complex chemical mixtures at different concentrations. The F-score, as a combined measure of recall and precision, was calculated using the ground truth data. The result was for centWave always higher than for matchedFilter and MZmine. The centWave algorithm is based on a sensitive detection of potentially interesting mass traces (ROIs), followed by an extensive chromatographic analysis, that reliably detects chromatographic peaks with different width via CWT. To allow for high sensitivity, baseline and noise are estimated locally. Some efforts are made to locate the exact chromatographic peak boundaries to provide accurate peak intensities. Feature quality can be assessed using numerous metrics, including signal to noise ratio, m/z fluctuation, and the residual of the Gaussian fit. Further development of the centWave algorithm will include an automatic estimation of the processing parameters.
In addition to centWave and the LC/MS data sets we have released the manual annotation of an LC/MS measurement of several pure compounds as a benchmark data set for both machine and software comparisons. The data sets are available at http://msbi.ipb-halle.de/msbi/centwave/.
CB performed the LC/MS measurements and was involved in the development of the centWave. RT designed and implemented the centWave algorithm. RT and SN performed the evaluation of the algorithms. All authors contributed to, read and approved the fnal manuscript.
Thanks to Colin Smith, who started the Bioconductor package XCMS and to Pan Du who wrote the MassSpecWavelet package for some of the CWT code.
- Oliver S, Winson M, Kell D, Baganz F: Systematic functional analysis of the yeast genome. Trends Biotechnol 1998, 16(9):373–378.View ArticlePubMed
- Fiehn O, Kopka J, Dörmann P, Altmann T, Trethewey R, Willmitzer L: Metabolite profiling for plant functional genomics. Nature Biotechnology 2000, 18: 115.View Article
- Dunn WB: Current trends and future requirements for the mass spectrometric investigation of microbial, mammalian and plant metabolomes. Physical Biology 2008, 5: 24. [http://stacks.iop.org/1478–3975/5/011001]View Article
- Roepenack-Lahaye Ev, Degenkolb T, Zerjeski M, Franz M, Roth U, Wessjohann L, Schmidt J, Scheel D, Clemens S: Profiling of Arabidopsis Secondary Metabolites by Capillary Liquid Chromatography Coupled to Electrospray Ionization Quadrupole Time-of-Flight Mass Spectrometry. Plant Physiology 2004, 134: 548–559.View Article
- Böttcher C, Roepenack-Lahaye Ev, Schmidt J, Schmotz C, Neumann S, Scheel D, Clemens S: Metabolome Analysis of Biosynthetic Mutants Reveals Diversity of Metabolic Changes and Allows Identification of a Large Number of New Compounds in Arabidopsis thaliana. Plant Physiol 2008, 108.117754. [http://www.plantphysiol.org/cgi/content/abstract/pp.108.117754v1]
- Tikunov Y, Lommen A, Vos Cd, Verhoeven H, Bino R, Hall R, Bovy A: A novel approach for nontargeted data analysis for metabolomics. Large-scale profiling of tomato fruit volatiles. Plant Physiol 2005, 139(3):1125–37.PubMed CentralView ArticlePubMed
- Smith C, Want E, O'Maille G, Abagyan R, Siuzdak G: XCMS: Processing mass spectrometry data for metabolite profiling using nonlinear peak alignment, matching and identification. Analytical Chemistry 2006, 78(3):779–787.View ArticlePubMed
- Katajamaa M, Oresic M: Processing methods for differential analysis of LC/MS profile data. BMC Bioinformatics 2005, 6: 179.PubMed CentralView ArticlePubMed
- Katajamaa M, Oresic M: Data processing for mass spectrometry-based metabolomics. 2007.
- Stolt R, Torgrip R, Lindberg J, Csenki L, Kolmert J, Schuppe-Koistinen I, Jacobsson S: Second-Order Peak Detection for Multicomponent High-Resolution LC/MS Data. Analytical Chemistry 2006, 78(4):975–983.View ArticlePubMed
- Aberg K, Torgrip R, Kolmert J, Schuppe-Koistinen I, Lindberg J: Feature detection and alignment of hyphenated chromatographic-mass spectrometric data Extraction of pure ion chromatograms using Kalman tracking. J Chromatogr A 2008, 1192: 139–146.View ArticlePubMed
- Lange E, Gröpl C, Reinert K, Kohlbacher O, Hildebrandt A: High-accuracy peak picking of proteomics data using wavelet techniques. Pacific Symposium on Biocomputing. Pacific Symposium on Biocomputing 2006, 243–254.
- Du P, Kibbe WA, Lin SM: Improved peak detection in mass spectrum by incorporating continuous wavelet transform-based pattern matching. Bioinformatics 2006, 22(17):2059–2065.View ArticlePubMed
- Conrad TOF, Leichtle A, Hagehulsmann A, Diederichs E, Baumann S, Thiery J, Schütte C: Beating the Noise: New Statistical Methods for Detecting Signals in MALDI-TOF Spectra Below Noise Level. In CompLife Edited by: Berthold MR, Glen RC, Fischer I. Lecture Notes in Computer Science Springer; 2006, 4216: 119–128. [http://dblp.uni-trier.de/db/conf/complife/complife2006.html#ConradLHDBTS06]
- McLerran DF, Feng Z, Semmes OJ, Cazares L, Randolph TW: Signal Detection in High-Resolution Mass Spectrometry Data. Journal of Proteome Research 2008, 7: 276–285.View ArticlePubMed
- Nordström A, O'Maille G, Qin C, Siuzdak G: Nonlinear data alignment for UPLC-MS and HPLC-MS based metabolomics: quantitative analysis of endogenous and exogenous metabolites in human serum. Anal Chem 2006, 78: 3289–3295.PubMed CentralView ArticlePubMed
- Danielsson R, Bylund D, Markides K: Matched filtering with background suppression for improved quality of base peak chromatograms and mass spectra in liquid chromatography – mass spectrometry. Analytica Chimica Acta 2002, (454):167–184.
- Andreev V, Rejtar T, Chen HS, Moskovets E, Ivanov A, Karger B: A Universal Denoising and Peak Picking Algorithm for LC-MS Based on Matched Filtration in the Chromatographic Time Domain. Analytical Chemistry 2003, 75(22):6314–6326.View ArticlePubMed
- Daubechies I: Ten lectures on wavelets. Philadelphia, PA, USA: Society for Industrial and Applied Mathematics; 1992.View Article
- Rijsbergen CJV: Information Retrieval. Newton, MA, USA: Butterworth-Heinemann; 1979.
- Yu H: Rmpi: Parallel Statistical Computing in R. R News 2002, 2(2):10–14. [http://CRAN.R-project.org/doc/Rnews/]
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.