Shape based kinetic outlier detection in realtime PCR
 Davide Sisti†^{1}Email author,
 Michele Guescini†^{2}Email author,
 Marco BL Rocchi^{1},
 Pasquale Tibollo^{2},
 Mario D'Atri^{1} and
 Vilberto Stocchi^{2}
DOI: 10.1186/1471210511186
© Sisti et al; licensee BioMed Central Ltd. 2010
Received: 11 December 2009
Accepted: 12 April 2010
Published: 12 April 2010
Abstract
Background
Realtime PCR has recently become the technique of choice for absolute and relative nucleic acid quantification. The gold standard quantification method in realtime PCR assumes that the compared samples have similar PCR efficiency. However, many factors present in biological samples affect PCR kinetic, confounding quantification analysis. In this work we propose a new strategy to detect outlier samples, called SOD.
Results
Richards function was fitted on fluorescence readings to parameterize the amplification curves. There was not a significant correlation between calculated amplification parameters (plateau, slope and ycoordinate of the inflection point) and the Log of input DNA demonstrating that this approach can be used to achieve a "fingerprint" for each amplification curve. To identify the outlier runs, the calculated parameters of each unknown sample were compared to those of the standard samples. When a significant underestimation of starting DNA molecules was found, due to the presence of biological inhibitors such as tannic acid, IgG or quercitin, SOD efficiently marked these amplification profiles as outliers. SOD was subsequently compared with KOD, the current approach based on PCR efficiency estimation. The data obtained showed that SOD was more sensitive than KOD, whereas SOD and KOD were equally specific.
Conclusion
Our results demonstrated, for the first time, that outlier detection can be based on amplification shape instead of PCR efficiency. SOD represents an improvement in realtime PCR analysis because it decreases the variance of data thus increasing the reliability of quantification.
Background
In the last few years, realtime quantitative polymerase chain reaction (realtime PCR) has become the technique of choice for absolute or relative quantification of gene expression due to its rapidity, accuracy and sensitivity [1–3]. Furthermore, recent advances in the sequencing of the human genome, mRNA and miRNA expression profiling of numerous cancer types, diseaseassociated polymorphism identification and the expanding availability of genomic sequence information for human pathogens have led to marked growth in molecular diagnostics [4–6].
The gold standard quantification method (Ct method) in realtime PCR assumes that the compared samples have similar PCR efficiencies. However, quantification by realtime PCR is very sensitive to slight differences in PCR efficiencies among samples. Indeed, a small difference of 5% in PCR efficiency will result in a threefold difference in the amount of DNA after 25 cycles of exponential amplification. Many factors present in samples as well as coextracted contaminants can inhibit PCR, confounding template amplification and analysis [7–10]. This is a major problem when working with biological samples. Severe inhibition will lead to falsenegative results, whereas a slight to moderate inhibition can result in an underestimation of the affected sample's DNA concentration [11]. Furthermore, amplification efficiency can fluctuate as a function of nonoptimal assay design, enzyme instability, or the presence of inhibitors [12]. Although a variety of methods have been developed to quantify template DNA [11, 13–17], very few allow simultaneous evaluation of template quantity and quality without the addition of an internal positive control that is coamplified with the target of interest. Hence Bar and coworkers proposed a method (called KOD) based on amplification efficiency calculation for the early detection of nonoptimal assay conditions [18, 19]. This approach is extremely straightforward and effective, but it is based on a PCR amplification efficiency calculation for which there is still not a method fully accepted by the scientific community. A large number of studies have attempted to calculate amplification efficiency assuming that PCR is inherently exponential in nature. Based on the assumption of the loglinearity region, constant amplification efficiency is calculated from the slope of linear regression in that window [20–23]. An alternative approach is based on the observation that PCR trajectory can be effectively modelled by the sigmoid function [14, 24] allowing PCR efficiency to be estimated using nonlinear regression fitting [15, 25, 26]. Recently, a simplified approach called "linear regression of efficiency" has allowed us to estimate amplification efficiency by applying linear regression analysis to the fluorescence readings within the central region of amplification profile [27]. Notably, it has been demonstrated that estimates of PCR efficiency vary widely according to the approach that has been adopted [28].
Very recently, Tichopad et al. [29] introduced a new quality control test for quantitative PCR; in this procedure the first derivative maximum and the second derivative maximum were estimated using a logistic fitting on the PCR trajectory. This approach allowed them to monitor the first half of the curve using two parameters.
Our study aims to develop a quality test tool, which is not based on amplification efficiency estimation, in order to detect samples that do not show an amplification kinetic similar to those of standard samples. In this work, a nonlinear fitting of Richards equation was used to parameterize PCR amplification profiles from a large sample set. The subsequent calculation of the variance of the estimated parameters and the development of a statistical measure based on the Mahalanobis distance allowed us to develop the SOD method (S hape based kinetic O utlier D etection). The SOD analysis of inhibited amplifications and the comparison of this method with KOD were investigated in detail.
Methods
Quantitative RealTime PCR
The DNA standard consisted of a pGEMT (Promega) plasmid containing a 104 bp fragment of the mitochondrial gene NADH dehydrogenase 1 (MTND1) as insert. This DNA fragment was produced by the ND1/ND2 primer pair (forward ND1: 5'ACGCCATAAAACTCTTCACCAAAG3' and reverse ND2: 5'TAGTAGAAGAGCGATGGTGAGAGCTA3'). This plasmid was purified using the Plasmid Midi Kit (Qiagen) according to the manufacturer's instructions. The final concentration of the standard plasmid was estimated spectophotometrically by averaging three replicate A_{260} absorbance determinations.
Realtime PCR amplifications were conducted using LightCycler^{®} 480 SYBR Green I Master (Roche) according to the manufacturer's instructions, with 500 nM primers and a variable amount of DNA standard in a 20 μ l final reaction volume. Thermocycling was conducted using a LightCycler^{®} 480 (Roche) initiated by a 10 min incubation at 95°C, followed by 40 cycles (95°C for 5 s; 60°C for 5 s; 72°C for 20 s) with a single fluorescent reading taken at the end of each cycle. Each reaction combination, namely starting DNA and inhibitor agent, was performed in triplicate and repeated in two separate amplification runs. All the runs were completed with a melt curve analysis to confirm the specificity of amplification and lack of primer dimers. Ct (fit point method) was determined by the LightCycler^{®} 480 software version 1.2 and exported into an MS Excel data sheet (Microsoft) for analysis after background subtraction (available as Additional file 1). For Ct (fit point method) evaluation, a fluorescence threshold manually set to 0.4 was used for all runs.
Estimation of PCR efficiency
The raw PCR data were used to calculate amplification efficiency. The PCR efficiency for each individual sample was derived from the slope of the regression line in the window of linearity [20]. Baseline correction and window of linearity identification were carried out using the latest version of LinRegPCR (v11.0) [23]. PCR efficiencies were estimated from four sample sets: standard amplification curves, standard amplification curves with the addition of tannic acid readouts, standard amplification curves with the addition of IgG readouts and standard amplification curves with the addition of quercitin readouts. The window of linearity calculated from all the data sets encompassed the fluorescence threshold of 0.4 chosen for the quantitative analysis.
Mathematical model of KOD
The mathematical model of KOD, based on efficiency, was proposed by Bar et al. [18]. Briefly, this was done comparing PCR efficiency of a sample (x_{ eff }) with the efficiencies of standard curve samples. A test sample is classified as an outlier if z > 1.96 with , where μ_{ eff }is the efficiency mean and σ_{ eff }is the standard deviation of the efficiency of standard curve samples. Alternatively, it is to be considered that the statistic is distributed as a χ^{2} with one degree of freedom; if χ^{2} > 3.84, we can reject the null hypothesis at α = 0.05.
Mathematical model of SOD
where x is the cycle number, F_{ x }is the reaction fluorescence at cycle x, F_{ max }is the maximal reaction fluorescence, F_{ b }is the background reaction fluorescence and b, c and d represents the estimated coefficients. Nonlinear regressions for 5parameter Richards functions were performed determining unweighted least squares estimates of parameters using the LevenbergMarquardt method.
The shape parameters used were the plateau value of amplification curve (F_{ max }), tangent straight line slope in inflection point (m) and ycoordinate of inflection point (Y_{ f }) (Additional file 2).
replacing Y_{ f }and F_{ max }, Eq. 5 can be simplified as: . In agreement with this equation the curve is symmetric (that is Asym = 0) when d = 1, or 2*Yf = Fmax. On the contrary, when d>1 we have 2*Yf<Fmax (the curve is asymmetric) hence Asym>0.
Statistical model of SOD
After developing a method to estimate three different shapeparameters (F_{ max }, Y_{ f }, m), the next step was to set a criterion to identify test samples that deviated from expected values. This was done using sample vector which can be calculated for each experimental amplification; if y belongs to a multivariate normal distribution, with mean vector and Σ the corresponding variancecovariance matrix, the (yμ)' Σ^{ 1 }(yμ) value (Mahalanobis distance) has asymptotic χ^{2} distribution, with 3 degrees of freedom. The Mahalanobis distance is based on correlations between variables through which different patterns can be identified and analyzed. It is a useful way of determining the similarity of an unknown multivariate sample set to a known one. It takes into account the correlations of the data set and is not dependent on the scale of measurements. Mean vector and variancecovariance matrix were calculated from shape parameters of standard curve samples. Then if χ^{2} > 7.81, we can reject the null hypothesis (with α = 0.05) and establish that the shape of the amplification curve is different from the shape of the standard curve samples, considering all three parameters [30]. All elaborations and graphics were obtained using Excel (Microsoft), Statistica 6.0 (Statsoft) and Statistical Package for Social Sciences (SPSS 13.0).
Results
Standard curve SOD analysis
The SOD model relies on the assumption that in order to achieve a reliable quantification, the amplification curves of unknown samples should not be significantly different from those of the standard curve. We introduced the idea that the amplification kinetic can be monitored by the shape of the amplification curve. The shape of amplification curves was parameterized using the nonlinear regression fitting of the Richards function on the fluorescence readings [11]. This mathematical procedure allowed us to obtain the five parameters characteristic of the Richards equation. These values were subsequently used to calculate the slope of the tangent at the inflection point (m), the ycoordinate of the inflection point (y_{ f }) and the maximum fluorescence value (F_{ max }) of the reading. Finally, these three parameters allowed us to create a "fingerprint" for each amplification curve.
OneSample KolmogorovSmirnov test of calibration curve.
Eff  y _{ f }  m  Fmax  

N = 72  
Mean (S.D.)  1.88 (0.02)  23.89 (2.86)  8.61 (1.20)  46.41 (6.07) 
KS value (p)  0.99 (0.28)  1.25 (0.09)  0.75 (0.63)  1.13 (0.15) 
In order to objectively define an amplification profile as an outlier, we introduced the variable Log(N_{ ob }/N_{ exp }), which estimates errors from quantification analysis using the Ct method. This variable relies on the residues estimated as the difference between calculated molecules, using the Ct method (Log of Number of Observed Molecules, referred to as LogN_{ ob }), and input DNA molecules (Log of Expected Molecules, referred to as LogN_{ exp }; in fact LogN_{ ob }LogN_{ exp }= Log(N_{ ob }/N_{ exp })). The ratio Log(N_{ ob }/N_{ exp }) showed a normal distribution satisfying the assumption of homoscedasticity (Additional file 4). It is thus possible to determine a 95% confidence interval (CI) for the variable Log(N_{ ob }/N_{ exp }). These residues showed a normal distribution regardless of the starting DNA template, with the average equal to zero and the standard deviation constant (σ = 0.041). In our database, out of a total of 72 runs used to construct the standard curve, 6 runs showed the ratio Log(N_{ ob }/N_{ exp }) out of the CI (Additional file 5). Subsequently, PCR efficiency (E_{ ff }) was also estimated for each amplification curve; the LinRegPCR software [20, 23] was used to fit the data points in the optimal range of the PCR exponential phase to obtain an automated evaluation of E_{ ff }(Table 1).
To determine how well outlier samples can be identified by KOD and SOD, we applied these statistical analyses to the runs of the standard curve; in particular we found that KOD identified 2 runs over the χ^{2} threshold value of 3.84 while SOD revealed 3 runs out of the CI (Additional file 5). These outliers are probably falsepositives due to the definition and intrinsic properties of the 95% CI.
Inhibitor effects on realtime amplification
These data led us to investigate the modifications of the parameters m, y_{ f }and F_{ max }in response to increasing inhibitor concentrations. Fig. 3C shows the increase in relative error of m, y_{ f }and F_{ max }in the presence of increasing tannic acid concentrations. Notably, these results also showed that curve asymmetry (Eq. 5) increased with higher inhibitor concentrations. This in turn demonstrates that not only the slope (m) and plateau (F_{ max }) of the curve decreased but also the shape changed moving towards a more and more Richards' type kinetic (Fig. 3D).
Furthermore, the asymmetry analysis showed an interesting singularity in the quercitin effects compared to those of tannic acid and IgG. In fact, quercitin led to kinetic alterations without a significant effect on the curve symmetry (Fig. 5D).
SOD versus KOD analysis
SOD and KOD analyses were used to identify samples with aberrant PCR kinetics, due to inhibitor presence, which might lead to erroneous quantifications. F_{ max }, m and y_{ f }values calculated from each amplification curve, obtained in the presence of increasing tannic acid, IgG or quercitin concentrations, were used to estimate the χ^{2}_{ SOD }value. Hence if the χ^{2}_{ SOD }value from an amplification curve was higher than the threshold value 7.81, the quantification was defined as an outlier. PCR efficiencies were also estimated and χ^{2}_{ KOD }values determined from the same amplifications. Quantification curves with a χ^{2}_{ KOD }values over 3.84 were rejected.
SOD and KOD analyses were also applied to realtime quantifications in the presence of IgG or quercitin as inhibitors. When amplification reactions were conducted in the presence of 20.5 mg/mL IgG, the suppression of amplification was efficiently revealed by both SOD and KOD, though SOD was more sensitive than KOD. In fact, SOD highlighted 17 outliers versus 15 revealed by KOD out of a total of 17 outliers (in the presence of IgG 17 runs led to a Log(N_{ ob }/N_{ exp }) out of 95% CI) (Fig. 6C and 6D). Analogous results were also obtained for quercitin. In the presence of 0.04 mg/mL of quercitin, SOD found 6 outliers compared to the 3 revealed by KOD out of a total of 6 outliers (Fig. 6E and 6F; for details of SOD and KOD analysis see Additional file 5).
Finally, we defined as true positives (TP) those amplifications showing χ^{2}>threshold value and those that led to a Log(N_{ ob }/N_{ exp }) ratio out of the 95% CI. Conversely, false positives (FP) were defined as samples that showed the χ^{2}>threshold value and a Log(N_{ ob }/N_{ exp }) ratio within the 95% CI. Consequently, true negatives (TP) were those amplifications showing χ^{2}<threshold value that led to a Log(N_{ ob }/N_{ exp }) ratio within the 95% CI and false negatives (FN) those showing χ^{2}<threshold value and Log(N_{ ob }/N_{ exp }) ratio out of the 95% CI.
Sensitivity and specificity of KOD and SOD analysis.
KOD  Tannic Acid  IgG  Quercitin 

Sensitivity  0.30***  0.76***  0.50*** 
Specificity  0.94  0.96  0.98 
SOD  Tannic Acid  IgG  Quercitin 
Sensitivity  0.93***  1.00***  1.00*** 
Specificity  1.00  0.94  0.98 
Discussion
A topic of great interest is the development of handfree tools for the detection of aberrant amplification profiles in realtime PCR analysis. Realtime PCR has rapidly become the most widely used technique in nucleic acid quantification. Although realtime PCR analysis has gained considerable attention in many fields of molecular biology, it is still troubled by significant technical problems [34]. Hence the present study has focused on the investigation of a new outlier detection approach which is not based on the PCR efficiency estimate but rather on the shape of the amplification profile.
The amplification nature of PCR makes it vulnerable to small differences in efficiencies of compared samples [20]. In fact, the current "gold standard" in realtime PCR analysis, the threshold cycle method (called Ct method), requires similar PCR efficiencies among compared samples.
However, dissimilarity in PCR efficiency results from different starting material sources, for example, different types of tissues [9]. Such differences might also be found when inhibitors of Taq DNA polymerase are present in cDNA samples [35] or in the presence of low quality SYBR green and/or dNTPs [36, 37]. Furthermore, the frequency of PCR inhibition [38] and different inhibitory effects even among replicates [39] highlight the need of kinetic quality assessment for each sample. Hence Bar et al. [18] proposed a statistical method, called KOD, to detect samples with dissimilar efficiencies.
KOD searches for outliers based on the main assumption that to obtain a reliable quantification, PCR runs have to show efficiencies which are not significantly different from each other. This condition is verified comparing the slopes of the straightline regression calculated in the windowoflinearity after the logtransformation of each readout fluorescence. In other words, if we return to raw data, the profile of the exponential curves in the windowoflinearity, mustn't be significantly different among compared runs. In the development of the SOD method we extended this concept to the whole curve, and all the runs included in the analysis have to show comparable amplification profiles.
The Ct method is based on the analysis of a serially diluted target. An example of this approach is presented in Fig. 1A careful examination of the obtained amplification profiles illustrates the central principle of the SOD method: all amplification curves are similar in shape and only the profile position is related to target quantity. The first amplification profiles, corresponding to the most concentrated samples, are found on the left, whereas samples with an increasing dilution factor regularly shift towards the right. This observation led us to the insight that an exclusion criterion could be based on the difference in shape rather than efficiency. This is in agreement with the work by Rutledge and Stewart [40] in which these authors described the amplification curve as a function of efficiency. Hence if efficiency determines the shape of a curve, by monitoring the shape of an amplification profile, information concerning the efficiency of amplification can be obtained.
Firstly, a "fingerprint" for each amplification curve using m, y_{ f }and F_{ max }resulting from the fitting of the Richards equation on raw data was obtained. Subsequently, these parameters were used to obtain the variancecovariance matrix in order to calculate the Mahalanobis distance [30]. This statistical measure is based on correlations among variables through which different patterns can be identified and analysed. In particular, the SOD analysis made use of the Mahalanobis distance to determine the similarity of an unknown sample compared to the standard set. This approach was very useful because it allowed us to evaluate not only the variance of single parameters (m, y_{ f }and F_{ max }), but also to quantify the reciprocal covariations among m, y_{ f }and F_{ max }.
F_{ max }was considered in the development of SOD because this parameter demonstrates successful amplification and usually, in suboptimal amplification conditions, the readouts do not reach characteristic F_{ max }values [9]. Examining our database, it was noted that F_{ max }showed high variance, thus it slightly affects χ^{2}_{ SOD }alone, but F_{ max }had a significant impact on the variancecovariance matrix. The parameter m describes the slope of the curve in the inflection point [11]. In our model, the higher the value of m, the higher the amplification rate is. However, this estimator does not directly indicate the amplification efficiency understood as the proportion between current and previous product amounts [38]. Finally, the asymmetry of amplification profiles was monitored by the relationship between F_{ max }and y_{ f }. It has been demonstrated that absolutely symmetrical PCR curves seldom occur, justifying the introduction of a fiveparameter fit [25]. Furthermore, in our previous work [11], it was demonstrated that the amplification reaction may deviate from a symmetric sigmoid curve to an asymmetric sigmoid (well described by Richards equation) in the presence of suboptimal efficiency. In fact, the goodness of fit of the logistic model progressively decreased with lower efficiency suggesting a change of PCR curve amplification shape [32].
The correlation analysis between m, y_{ f }and F_{ max }obtained from the standard curve and input DNA demonstrated that these shape parameters are concentrationindependent. This supports our experimental hypothesis that all the amplification curves of the standard curve are similar in shape and only the profile position determines target quantity. In the presence of PCR inhibition, it was found that increasing concentrations of tannic acid and IgG resulted in decreasing F_{ max }and m values, while asymmetry increased with higher inhibitor concentrations (when asymmetry increases, y_{ f }decreases more than the corresponding F_{ max }; Fig. 3 and 4). It may be that tannic acid inhibition is simply due to fluorescence quenching since we found a dramatic decrease in F_{ max }and a slide curve slope decrease. However, we also showed that fluorescence asymmetry increased demonstrating that tannic acid produced an amplification kinetic distortion. The addition of quercitin to PCR amplifications produced very interesting data. In fact, we found decreased F_{ max }and m values in the presence of high inhibitor concentrations, however this flavonid did not induce an asymmetric modification of the curves (Fig. 5D). The reported data clearly demonstrate that the SOD method can identify nonoptimal PCR kinetics resulting from different inhibition models. Furthermore, the results obtained in the presence of quercitin highlight the importance of using a multivariate approach.
When comparing SOD to KOD performance, it was found that SOD was more sensitive than KOD in all the tested settings. SOD and KOD were equally specific in the presence of IgG and quercitin, whereas SOD was more specific than KOD in the presence of tannic acid.
Furthermore, the SOD method presents several advantages over KOD; SOD is completely handfree. Indeed, it is not necessary for the user to identify a window of analysis as in the KOD method, and more importantly, SOD does not rely on a constant efficiency value avoiding all the problems connected with its determination [28, 40, 41]. As previously reported, variable PCR efficiency determination can lead to different results contributing to erroneous and spread quantifications [19]. Moreover, logtransformation of fluorescence data that could be responsible for bias in the analysis are avoided.
The SOD method has been developed for the chemistry Sybr Green, and the application of this procedure to other chemistries such as TaqMan, needs to be evaluated extensively.
Very recently, Tichopad et al. [29] proposed a new KOD procedure based on Malahanobis statistic [30]. In this study the first derivative maximum and the second derivative maximum were estimated using a logistic fitting on the central portion of the PCR trajectory. Using these two parameters these authors proposed monitoring only the first half of the curve. On the contrary, the SOD method is based on the possibility of describing the whole PCR trajectory using Richards equation. SOD represents a continuation and an extension of the application of Richards equation to realtime PCR readings [11]. We think that the SOD method introduces original concepts that are not found in the recently developed method described by Tichopad et al. [29]. SOD takes advantage of the possibility of describing the shape of the whole PCR trajectory through the combination of the parameters m, y_{ f }and F_{ max }while the method by Tichopad et al. [29] focuses on two key points of the trajectory: the maximum of the first and second derivative. Furthermore, in the SOD method we used quite a different metric approach. Although other multivariate methods are available for similar tasks (support vector machines, Kmeans cluster), we used asymptotic distribution of the Mahalanobis distance because it is a logical extension of the KOD method, which is based on univariate normal distribution.
Conclusion
We demonstrated for the first time that a comparison of the shape variation of an amplification profile with the shape of standard profiles can be used to exclude aberrant samples from Ct analysis. This allows us to avoid the spread of results and therefore increases the potential of quantification analysis.
Hence we propose SOD as a handfree quality control method in realtime PCR analysis with applications in any field of molecular diagnostics.
Notes
Abbreviations
 Ct:

threshold cycle
 IgG:

immunoglobulin G
 SOD:

shape based kinetic outlier detection
 KOD:

kinetic outlier detection
 Asym:

Asymmetry.
Declarations
Authors’ Affiliations
References
 Gingeras TR, Higuchi R, Kricka LJ, Lo YM, Wittwer CT: Fifty years of molecular (DNA/RNA) diagnostics. Clin Chem 2005, 51(3):661–671. 10.1373/clinchem.2004.045336View ArticlePubMedGoogle Scholar
 Nolan T, Hands RE, Bustin SA: Quantification of mRNA using realtime RTPCR. Nature Protocols 2006, 1(3):1559–1582. 10.1038/nprot.2006.236View ArticlePubMedGoogle Scholar
 VanGuilder HD, Vrana KE, Freeman WM: Twentyfive years of quantitative PCR for gene expression analysis. Bio Techniques 2008, 44(5):619–626.Google Scholar
 Gunson RN, Bennett S, Maclean A, Carman WF: Using multiplex real time PCR in order to streamline a routine diagnostic service. J Clin Virol 2008, 43(4):372–375. 10.1016/j.jcv.2008.08.020View ArticlePubMedGoogle Scholar
 Watzinger F, Ebner K, Lion T: Detection and monitoring of virus infections by realtime PCR. Molecular aspects of medicine 2006, 27(2–3):254–298. 10.1016/j.mam.2005.12.001View ArticlePubMedGoogle Scholar
 Kaltenboeck B, Wang C: Advances in realtime PCR: application to clinical laboratory diagnostics. Advances in clinical chemistry 2005, 40: 219–259. full_textView ArticlePubMedGoogle Scholar
 Akane A, Matsubara K, Nakamura H, Takahashi S, Kimura K: Identification of the heme compound copurified with deoxyribonucleic acid (DNA) from bloodstains, a major inhibitor of polymerase chain reaction (PCR) amplification. Journal of forensic sciences 1994, 39(2):362–372.View ArticlePubMedGoogle Scholar
 Wilson IG: Inhibition and facilitation of nucleic acid amplification. Applied and environmental microbiology 1997, 63(10):3741–3751.PubMedPubMed CentralGoogle Scholar
 Tichopad A, Didier A, Pfaffl MW: Inhibition of realtime RTPCR quantification due to tissuespecific contaminants. Mol Cell Probes 2004, 18(1):45–50. 10.1016/j.mcp.2003.09.001View ArticlePubMedGoogle Scholar
 Rossen L, Norskov P, Holmstrom K, Rasmussen OF: Inhibition of PCR by components of food samples, microbial diagnostic assays and DNAextraction solutions. International journal of food microbiology 1992, 17(1):37–45. 10.1016/01681605(92)90017WView ArticlePubMedGoogle Scholar
 Guescini M, Sisti D, Rocchi MB, Stocchi L, Stocchi V: A new realtime PCR method to overcome significant quantitative inaccuracy due to slight amplification inhibition. BMC bioinformatics 2008, 9: 326. 10.1186/147121059326View ArticlePubMedPubMed CentralGoogle Scholar
 Kainz P: The PCR plateau phase  towards an understanding of its limitations. Biochimica et biophysica acta 2000, 1494(1–2):23–27.View ArticlePubMedGoogle Scholar
 Livak KJ, Schmittgen TD: Analysis of relative gene expression data using realtime quantitative PCR and the 2(Delta Delta C(T)) Method. Methods (San Diego, Calif) 2001, 25(4):402–408.View ArticleGoogle Scholar
 Liu W, Saint DA: Validation of a quantitative method for real time PCR kinetics. Biochem Biophys Res Commun 2002, 294(2):347–353. 10.1016/S0006291X(02)004783View ArticlePubMedGoogle Scholar
 Rutledge RG: Sigmoidal curvefitting redefines quantitative realtime PCR with the prospective of developing automated highthroughput applications. Nucleic acids research 2004, 32(22):e178. 10.1093/nar/gnh177View ArticlePubMedPubMed CentralGoogle Scholar
 Pfaffl MW: A new mathematical model for relative quantification in realtime RTPCR. Nucleic acids research 2001, 29(9):e45. 10.1093/nar/29.9.e45View ArticlePubMedPubMed CentralGoogle Scholar
 Goll R, Olsen T, Cui G, Florholmen J: Evaluation of absolute quantitation by nonlinear regression in probebased realtime PCR. BMC bioinformatics 2006, 7: 107. 10.1186/147121057107View ArticlePubMedPubMed CentralGoogle Scholar
 Bar T, Stahlberg A, Muszta A, Kubista M: Kinetic Outlier Detection (KOD) in realtime PCR. Nucleic acids research 2003, 31(17):e105. 10.1093/nar/gng106View ArticlePubMedPubMed CentralGoogle Scholar
 Kontanis EJ, Reed FA: Evaluation of realtime PCR amplification efficiencies to detect PCR inhibitors. Journal of forensic sciences 2006, 51(4):795–804. 10.1111/j.15564029.2006.00182.xView ArticlePubMedGoogle Scholar
 Ramakers C, Ruijter JM, Deprez RH, Moorman AF: Assumptionfree analysis of quantitative realtime polymerase chain reaction (PCR) data. Neurosci Lett 2003, 339(1):62–66. 10.1016/S03043940(02)014234View ArticlePubMedGoogle Scholar
 Wilhelm J, Pingoud A, Hahn M: Validation of an algorithm for automatic quantification of nucleic acid copy numbers by realtime polymerase chain reaction. Anal Biochem 2003, 317(2):218–225. 10.1016/S00032697(03)001672View ArticlePubMedGoogle Scholar
 Wilhelm J, Pingoud A, Hahn M: SoFAR: software for fully automatic evaluation of realtime PCR data. Bio Techniques 2003, 34(2):324–332.Google Scholar
 Ruijter JM, Ramakers C, Hoogaars WM, Karlen Y, Bakker O, Hoff MJ, Moorman AF: Amplification efficiency: linking baseline and bias in the analysis of quantitative PCR data. Nucleic acids research 2009, 37(6):e45. 10.1093/nar/gkp045View ArticlePubMedPubMed CentralGoogle Scholar
 Liu W, Saint DA: A new quantitative method of real time reverse transcription polymerase chain reaction assay based on simulation of polymerase chain reaction kinetics. Anal Biochem 2002, 302(1):52–59. 10.1006/abio.2001.5530View ArticlePubMedGoogle Scholar
 Spiess AN, Feig C, Ritz C: Highly accurate sigmoidal fitting of realtime PCR data by introducing a parameter for asymmetry. BMC bioinformatics 2008, 9: 221. 10.1186/147121059221View ArticlePubMedPubMed CentralGoogle Scholar
 Qiu H, Durand K, RabinovitchChable H, Rigaud M, Gazaille V, Clavere P, Sturtz FG: Gene expression of HIF1alpha and XRCC4 measured in human samples by realtime RTPCR using the sigmoidal curvefitting method. Bio Techniques 2007, 42(3):355–362.Google Scholar
 Rutledge RG, Stewart D: A kineticbased sigmoidal model for the polymerase chain reaction and its application to highcapacity absolute quantitative realtime PCR. BMC biotechnology 2008, 8: 47. 10.1186/14726750847View ArticlePubMedPubMed CentralGoogle Scholar
 Cikos S, Bukovska A, Koppel J: Relative quantification of mRNA: comparison of methods currently used for realtime PCR data analysis. BMC molecular biology 2007, 8: 113. 10.1186/147121998113View ArticlePubMedPubMed CentralGoogle Scholar
 Tichopad A, Bar T, Pecen L, Kitchen RR, Kubista M, Pfaffl MW: Quality control for quantitative PCR based on amplification compatibility test. Methods 2010, 50(4):308–312. 10.1016/j.ymeth.2010.01.028View ArticlePubMedGoogle Scholar
 Rencher AC: Methods of Multivariate Analysis. 2nd edition. Wiley, Printed in US; 2002.View ArticleGoogle Scholar
 Young CC, Burghoff RL, Keim LG, MinakBernero V, Lute JR, Hinton SM: PolyvinylpyrrolidoneAgarose Gel Electrophoresis Purification of Polymerase Chain ReactionAmplifiable DNA from Soils. Applied and environmental microbiology 1993, 59(6):1972–1974.PubMedPubMed CentralGoogle Scholar
 Tichopad A, Polster J, Pecen L, Pfaffl MW: Model of inhibition of Thermus aquaticus polymerase and Moloney murine leukemia virus reverse transcriptase by tea polyphenols (+)catechin and ()epigallocatechin3gallate. J Ethnopharmacol 2005, 99(2):221–227. 10.1016/j.jep.2005.02.021View ArticlePubMedGoogle Scholar
 Nolan T, Hands RE, Ogunkolade W, Bustin SA: SPUD: a quantitative PCR assay for the detection of inhibitors in nucleic acid preparations. Anal Biochem 2006, 351(2):308–310. 10.1016/j.ab.2006.01.051View ArticlePubMedGoogle Scholar
 Murphy J, Bustin SA: Reliability of realtime reversetranscription PCR in clinical diagnostics: gold standard or substandard? Expert review of molecular diagnostics 2009, 9(2):187–197. 10.1586/14737159.9.2.187View ArticlePubMedGoogle Scholar
 Chandler DP, Wagnon CA, Bolton H Jr: Reverse transcriptase (RT) inhibition of PCR at low concentrations of template and its implications for quantitative RTPCR. Applied and environmental microbiology 1998, 64(2):669–677.PubMedPubMed CentralGoogle Scholar
 Kubista M, Stahlberg A, Bar T: Lightup probe based realtime QPCR. In Genomics and Proteomics Technologies Proceedings of SPIE Edited by: TW Raghavachari R. 2001, 53–58.View ArticleGoogle Scholar
 Karsai A, Muller S, Platz S, Hauser MT: Evaluation of a homemade SYBR green I reaction mixture for realtime PCR quantification of gene expression. Bio Techniques 2002, 32(4):790–792. 794–796 794796Google Scholar
 Tichopad A, Dzidic A, Pfaffl MW: Improving quantitative realtime RTPCR reproducibility by boosting primerlinked amplification efficiency. Biotechnology Letters 2002, 24: 2053–2056. 10.1023/A:1021319421153View ArticleGoogle Scholar
 Rosenstraus M, Wang Z, Chang SY, DeBonville D, Spadoro JP: An internal control for routine diagnostic PCR: design, properties, and effect on clinical performance. Journal of clinical microbiology 1998, 36(1):191–197.PubMedPubMed CentralGoogle Scholar
 Rutledge RG, Stewart D: Critical evaluation of methods used to determine amplification efficiency refutes the exponential character of realtime PCR. BMC molecular biology 2008, 9: 96. 10.1186/14712199996View ArticlePubMedPubMed CentralGoogle Scholar
 Skern R, Frost P, Nilsen F: Relative transcript quantification by quantitative PCR: roughly right or precisely wrong? BMC molecular biology 2005, 6(1):10. 10.1186/14712199610View ArticlePubMedPubMed CentralGoogle Scholar
Copyright
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.