 Methodology
 Open access
 Published:
Compressed sensing improved iterative reconstructionreprojection algorithm for electron tomography
BMC Bioinformatics volume 21, Article number: 202 (2020)
Abstract
Background
Electron tomography (ET) is an important technique for the study of complex biological structures and their functions. Electron tomography reconstructs the interior of a threedimensional object from its projections at different orientations. However, due to the instrument limitation, the angular tilt range of the projections is limited within +70^{∘} to −70^{∘}. The missing angle range is known as the missing wedge and will cause artifacts.
Results
In this paper, we proposed a novel algorithm, compressed sensing improved iterative reconstructionreprojection (CSIIRR), which follows the schedule of improved iterative reconstructionreprojection but further considers the sparsity of the biological ultrastructural content in specimen. The proposed algorithm keeps both the merits of the improved iterative reconstructionreprojection (IIRR) and compressed sensing, resulting in an estimation of the electron tomography with faster execution speed and better reconstruction result. A comprehensive experiment has been carried out, in which CSIIRR was challenged on both simulated and realworld datasets as well as compared with a number of classical methods. The experimental results prove the effectiveness and efficiency of CSIIRR, and further show its advantages over the other methods.
Conclusions
The proposed algorithm has an obvious advance in the suppression of missing wedge effects and the restoration of missing information, which provides an option to the structural biologist for clear and accurate tomographic reconstruction.
Background
Electron tomography (ET) is an important technique for the study of complex biological structures and their functions, which reconstructs the interior of a threedimensional object from its projections at different orientations [1]. However, due to the instrument limitation, the angular tilt range of the projections is limited within +70^{∘} to −70^{∘}. Such a series of the projections compose an anglelimited projection dataset, in which the unsampled high tilt angles are called the missing wedge and will cause artifacts. Therefore, the reconstruction of the tomogram in ET is an optimization problem with incomplete information. Additionally, the optimization problem becomes more difficult due to the extreme noise caused by the low electron doses in the collection of Electron Microscopy (EM) images.
Traditional 3D reconstruction methods, such as Weighted backprojection (WBP) and Filtered backprojection (FBP), suffer from the incomplete data and extremely noisy projections, resulting in the undesired artifacts. Many algorithms have been proposed to deal with the incompleteness caused by the limited projections and strong noise. By combining the algebra reconstruction technique (ART) and the nonlinear diffusion (ND) filter technique, filtered iterative reconstruction technique (FIRT) [2] could partially restore some information at the nonsampled angular region. Iterative reconstructionreprojection (IRR) [3–5] and its variations, such as discrete IRR algorithm and the finite impulse response (FIR) lowpass filter (DIRRLF) [6], iterative reconstructionreprojection (IRR) algorithm with total variation (TV) constraint (IRRTV) [7], iterative reconstructionreprojection in projection space (IRRP) algorithm [8] and algebraic iterative reconstructionreprojection (AIRR) [9], estimate the missing projects by an iterative optimization procedure between the projection and image domain.
Improved iterative reconstructionreprojection (IIRR) [10, 11] is one of the variations of the IRR algorithm, whose convergence under extremely noisy condition has been proved [12]. However, without prior information, IIRR is not efficient in suppressing the effect of the missing wedge and restoring the missing information for ET data, because of the low correlation between known projections and unknown projections in the missing wedge. On the other hand, other algorithms reduce artifacts by introducing prior information. Iterative Compressedsensing Optimized Nonuniform fast Fourier transform reconstruction (ICON) [13–15] combines compressed sensing (CS) with nonuniform fast Fourier transform (NUFFT) in order to restore the missing information. There are two main steps in ICON. One is the fidelity preservation step which minimizes the difference between the reconstructed volume and the projections. Another one is prior sparsity restriction step which increases the sparsity of the reconstructed volume. Nonetheless, a drawback of ICON is its slow computational speed.
In this paper, we proposed a novel algorithm which follows the schedule of IIRR but further considers the sparsity of the biological ultrastructural content in specimen, resulting in an estimation of the electron tomography with faster execution speed and better reconstruction result. A comprehensive experiment on both the simulated and realworld dataset is carried out to estimate the performance of our algorithm and compare the new algorithm with other stateoftheart methods. The experimental results indicate that the proposed algorithm has the ability to suppress the effect of missing wedge and restore the missing information within the missing wedge. Specifically, compared with the classical methods, our algorithm can achieve better reconstruction with faster convergence.
Preliminaries
Without loss of generality, our following discussion is defined in twodimensional space. Nevertheless, the algorithm we proposed can be extended to threedimensional space easily.
From filtered backprojection (FBP) to iterative reconstructionreprojection (IRR)
The filtered backprojection (FBP) algorithm is one of the oldest algorithms in computing tomography. Its mathematical model could be concluded as following:
Given a function f(x) (\(x\in \mathbb {R}^{2}\)), the projections of function f is defined as a linear integral along a straight line \(L=\{x\in \mathbb {R}^{2}\left (\cos \alpha \right) x_{1}+\left (\sin \alpha \right) x_{2}=s\}\)
where \(\mathscr {R}\) is the Radon transform operator, \(s\in \mathbb {R}\) is the distance of line L from the origin, α is the angle between normal vector to L and the x_{1} axis. Suppose the value of p on \(\Omega \subset \{\left (s,\alpha \right)\alpha \in \left (0,\pi \right ],s\in \mathbb {R}\}\) is known in prior. Let \(\chi _{\Omega }(s,\alpha)=\left \{ \begin {array}{c} 1,(s,\alpha)\in \Omega \\ 0,(s,\alpha)\notin \Omega \end {array}\right.\) be the characteristic function of set Ω,p_{unknown}(s,α)=(1−χ_{Ω}(s,α))p(s,α) be the unknown projections and p_{known}(s,α)=χ_{Ω}(s,α)p(s,α) be the known projections. FBP will first apply a filter on the projections and then backproject these filtered projections to get an estimation of the original signal f:
1) Convolution (Filtering)
where * denotes convolution, \(\mathscr {F}_{1}^{1}\) denotes the one dimensional Fourier transform.
2) Backprojection
where \(\widetilde {f}\) is the estimation of f, x_{i} is the ith element of x, \(\mathscr {B}\) is the backprojection operator. It should be noted that, in FBP, the unknown projections p_{unknown} have been set to 0.
Meanwhile, the iterative reconstructionreprojection algorithm (IRR) is an iterative reconstruction method which is able to suppress the effect of insufficient tilt angles by estimating the unknown projections. At the beginning, all the unknown projections in the system \(p_{unknown}^{1}\) are set to 0 as in FBP. During the computational process of IRR, for the ith iteration, the ith estimation of f will be reconstructed from both the known projections p_{known} and the ith estimated unknown projections \(p_{unknown}^{i}\):
However, indeed, we do not know the unknown projections \(p_{unknown}^{i}\). In IRR, the (i)th unknown projections are estimated from f in the previous iteration:
By estimating the unknown projections rather than setting them to 0, IRR suffers less from the ’missing wedge’ problem than FBP.
The improved algorithms based on IRR
Improved iterative reconstructionreprojection (IIRR)
A drawback of IRR is that the convergence of IRR under extremely noisy condition is not guaranteed. By introducing a relaxed parameter λ into IRR, the convergence under extremely noisy condition has been proved. More precisely, IIRR changes Eq. (4) into
where 0<λ<1 is a relaxed parameter. The smaller value of λ considers more in the known projections while larger value of λ considers more about the unknown projections. IIRR can restrain the artifact caused by the incomplete data in Computerized Tomography (CT).
Iterative reconstructionreprojection with total variation (IRRTV)
IRRTV is an reconstruction method which is developed for fewview projections, in which the angular tilt range of the projections is not limited but the number of projections is limited. IRRTV poses the reconstruction problem as an optimization problem with the form as:
where Af=b is the fidelity term,
is the total variation norm, f is the unknown 3D reconstruction volume, \(\left (x_{1},x_{2},x_{3}\right)\in \mathbb {Z}^{3}\) is the coordinate of voxels in unknown volume, A is the projection matrix, and b is the observations or projections. By introducing TV into IRR, IRRTV achieves an accelerated convergence rate compared with the classic IRR algorithm.
Specifically, in each iteration before the estimation of unknown projections, IRRTV minimizes the TV norm by deepest gradient descent:
where d is step size, f^{i} is the ith estimation of f, and TV(·) is the TV norm.
Matching pursuit
Matching pursuit (MP) is an algorithm which represents a signal in a redundant basis. Let \(\mathscr {D}=\{g_{\gamma }\}_{\gamma \in \Gamma }\) be a family of vectors in Hilbert space and f be a signal. f can be represented as a linear combination of N vectors selected from \(\mathscr {D}\)
where \(a_{\gamma _{n}}\) is the weight of \(g_{\gamma _{n}}, g_{\gamma _{n}}\) is the vectors selected from \(\mathscr {D}\), N is the total number of vectors selected from \(\mathscr {D}\). The basic idea of MP is choosing vectors from \(\mathscr {D}\) greedily. More precisely, let R_{1}=f be the initial approximation error. For i=1,2,3,..., MP finds \(g_{\gamma _{i}}\) with maximum absolute value of inner product \(\phantom {\dot {i}\!}<R_{i},g_{\gamma _{i}}>\), where \(\phantom {\dot {i}\!}R_{i}=f\sum _{n=1}^{i}a_{\gamma _{n}}g_{\gamma _{n}}\) is the ith approximation error. Then we set \(\phantom {\dot {i}\!}a_{i}=<R_{i},g_{\gamma _{i}}>\). The process continues until the approximation error is small enough.
Methods
Problem definition
Despite the success of IIRR in Computed Tomography (CT), IIRR is not efficient in suppressing the effect of the missing wedge and restore the missing information for ET data. Here, an algorithm based on the schedule of IIRR with the consideration of sparse information is proposed to achieve better ET reconstruction.
According to the previous study [13], the biological ultrastructural content is relatively sparse with respect to the surrounding solvent. Such sparsity is used as the prior information in our proposed method. Formally, let f(x) be the space function (\(x\in \mathbb {R}^{2}\)) representing the sample and suppose that f is supported on set T which is a subset of \(\mathbb {Z}^{2}\). The consequent optimization problem could be defined as
where \(\Vert f\Vert _{0}=\#\{x\in \mathbb {Z}^{2}\cap Tf(x)\neq 0\}\) is the l_{0} norm of f, x is a 2 dimensional vector representing the coordinate of pixels in picture f, \(\mathscr {R}\) is the Radon transform, p_{known} is the known projections, χ_{Ω} is the characteristic function of set Ω, and Ω is the subset of projection space corresponding to the known projections.
Compressed sensing improved iterative reconstructionreprojection (CSIIRR)
We designed a new algorithm to solve the optimization problem (11) by combining IIRR and CS. The proposed algorithm is elaborated as follows.

Step 0: Preprocess

Step 1: Estimate the unknown projections

Step 2: Reconstruct via modified matching pursuit (MMP), return to step 1
Step 0: preprocess
Alignment is needed for the raw projections before reconstruction for ET data [16, 17]. Then we need an initial guess of specimen f which can be the reconstruction result via any reconstruction method or even a zero function. We denote the initial guess of f as f^{0}.
Step 1: estimate the unknown projections
Let f^{k−1} be the reconstruction result in the (k−1)th step, k=1,2,.... We estimate the unknown projections by reprojecting f^{k−1} at the corresponding tilt angles. We denote the estimated projections as
where the notation \(\mathscr {R}\) denotes the Radon transform, Ω is the subset of projection space corresponding to the known projections, and χ_{Ω} denotes the characteristic function of Ω.
Step 2: reconstruct via modified MP (MMP)
In this step, we want to reconstruct specimen f^{k} from the known projections p_{known} and the estimated projections \(p_{unknown}^{k1}\). We seek for the sparsest f^{k} such that \(\mathscr {R}f^{k1}=p_{known}+\lambda p_{unknown}^{k1}\), where 0<λ<1 is a relaxed parameter. A small λ is suggested for extremely noisy data because the reliability of estimated unknown projections is low with high noise levels. So we introduced compressed sensing to this step by a modified matching pursuit method. The detailed process of our modified matching pursuit method is described in Algorithm 1.
As shown in Algorithm 1, we first initialize the approximation of reconstruction result \(f_{mmp}^{0}\) to zero vector, and the residual R_{0} to \(p_{known}+\lambda p_{unknown}^{k1}\). In the lth iteration, we want to find the largest M absolute values of pixel value in \(\chi _{T}\mathscr {B}(R_{l1}*\mathscr {F}^{1}\omega )\) and restore the location of the M voxels in x_{1},x_{2},...,x_{M}. The parameter M affects the sparsity of the reconstruction result. A small M is suggested if the biological ultrastructural content is sparse in the specimen. Next, we use the restored information to update the approximation \(f_{mmp}^{l}\) by Eq. (??) and the consequent residual R_{l} by Eq. (??). The process continues until ∥R∥<ε. After obtaining f^{k}, the algorithm returns to step 1 in order to optimize the estimated projections \(p_{unknown}^{k}\).
The main advantage of combining CS with IIRR is to introduce sparsity into IIRR. There are two main differences between the Algorithm 1 and MP methods. First, Algorithm 1 estimates the tomogram not only from the known projections p_{known} but also from the estimated projections \(p_{unknown}^{k1}\), which makes the proposed algorithm more stable than MP. Second, instead of choosing only one atom, Algorithm 1 chooses multiple pixels in each iteration. By doing so, Algorithm 1 can achieve a quick convergence compared with the classic MP. Moreover, there are also some differences between IRRTV and CSIIRR. First, the two methods choose different regularization. IRRTV minimizes the TV norm while CSIIRR minimizes the l_{0} norm. Second, IRRTV adds a ’deepest gradient descent’ step after the back projection, which is timeconsuming. Third, IRRTV focuses on the fewview problem in CT while CSIIRR focuses on the anglelimited problem in ET. Especially, CSIIRR is able to handle extremely noisy data on which IRRTV usually fails.
Results
The CSIIRR is challenged by both the simulated data and realworld data. Expect for CSIIRR, the filtered backprojection (FBP), IIRR, weighted backprojection (WBP), Iterative Compressedsensing Optimized Nonuniform fast Fourier transform reconstruction (ICON) [13, 14] and total variation (TV) are also tested in our experiment as a comparison. (Here, the algorithms of FBP, IIRR and CSIIRR are implemented with MATLAB, the implementation of ICON is download from http://ear.ict.ac.cn and the implementation of WBP comes from the ’tilt’ module in IMOD.) In the experiment, the relaxation factor λ is set to 0.99 for both IIRR and the CSIIRR.
Performance on simulated data
Overall evaluation
First, CSIIRR is challenged on the noisefree simulated data. The ribosome structure downloaded from EMDB3489 [18] is used as the groundtruth, whose volume size of EMDB3489 is 400^{3} voxels. A set of the projections with the range of the tilt angle from 60^{∘} to −60^{∘} is generated as the measured observations, with an angular increment of 1^{∘}.
To reconstruct the tomogram, the number of iterations is set to 10 in IIRR. For the sake of fairness, the number of outerloop iterations in our proposed algorithm is also set to 10. To make reconstruction results converge, the number of iterations for modified MP in the proposed algorithm is set to 50. To trade off the convergence rate of CSIIRR and the sparsity of the output, the number of atoms selected in each iteration (M in Algorithm 1) is set to 335103 in modified MP.
Figure 1 demonstrates the reconstruction results obtained by FBP, IIRR, CSIIRR, ICON and TV. As shown in the first row, the central slice of the reconstruction produced by CSIIRR suffers the least artifacts caused by the incompleteness of the sampling angle compared with the ones of other methods. The logscaled power spectrums on the second row of Fig. 1 shows that CSIIRR has filled some missing data in the missing wedge. Furthermore, the phase difference of the tomograms reconstructed by different methods has been presented in the third row of Fig. 1 (The pointtopoint phase difference [13] Δ of a and b is defined as \(\Delta =\vert \frac {\vert ab\vert }{\pi }1\vert \).), which shows that CSIIRR has restored some correct information in the missing wedge. Moreover, the histograms demonstrated in the fourth row of Fig. 1 further support the conclusion made from the logscaled power spectrum, in which the histogram shows that the reconstruction result by CSIIRR has a much more similar voxel distribution with the groundtruth, compared with the ones obtained from other methods. Additionally, the histogram accurately presents the sparseness of the reconstruction generated by CSIIRR.
Convergence analysis
Except for the reconstruction accuracy, the convergence and running speed are also the important issues for a novel iterative method. Here, we first investigated the convergence of the modified MP algorithm (MMP) and the classic matching pursuit algorithm (MP) under different parameter configuration. And then compared the runtime of the proposed CSIIRR with the other classical methods (e.g. WBP, IIRR and ICON). The simulated data shown in the previous section is used as the benchmark dataset. When comparing the proposed modified MP and classic MP, we omitted the estimation of unknown projection in MMP (i.e., we set \(p_{unknown}^{k1}=0\) in Algorithm 1) for fair comparison. One point should be noted is that the classic MP is a special case of MMP if setting the parameter M to 1. The Pearson correlation coefficient (PCC) between the groundtruth and each turn’s reconstruction result is measured to indicate the improvement of reconstruction quality during the iteration process. Figure 2 demonstrates the change of PCC value for the two methods. It can be found that the MMP algorithm has a much faster convergence rate compared with the classic MP algorithm, no matter how we set the parameters. Furthermore, in the early iterations, the larger the M is, the faster the convergence speed will be.
When comparing the runtime between the whole CSIIRR framework and the classic WBP, IIRR, and ICON methods, the M used in modified MP is set to 335103 (i.e., with 0.01 ratio). We use structural similarity (SSIM) which is a measurement of image quality to evaluate CSIIRR. Figure 3 shows the experimental result, in which the SSIM curves of CSIIRR, ICON, WBP and IIRR with respect to the running time are demonstrated. It can be observed that CSIIRR converged to the highest SSIM value with the fastest convergence rate.
Effect of tilt angle change
Next, we tested the robustness of CSIIRR on different tilt angle ranges and intervals.
CSIIRR was tested on simulated data with angular range set to −50^{∘} to 50^{∘},−60^{∘} to 60^{∘} and −70^{∘} to 70^{∘} (Fig. 4). As shown in Fig. 4, CSIIRR’s FSC curve is significantly higher than the ones of other methods.
Furthermore, CSIIRR was tested on simulated data with angular intervals 1^{∘},2^{∘} and 3^{∘} (Fig. 5). As shown in Fig. 5, CSIIRR’s FSC curve has significantly higher coefficients than that of other methods, especially for high frequency.
Effect of noise
The effect of noise on the behavior CSIIRR has been investigated. The noise with signaltonoise ratio SNR=0.5,1,2 is added to the projections and the tomogram is reconstructed by WBP, IIRR, ICON and CSIIRR. The Fourier shell correlation (FSC) between the reconstructed tomogram and the groundtruth is calculated to show the reconstruction quality of each method. Figure 6 shows corresponding FSC curve. For the data with SNR=0.5 and 1(Fig. 6a, b), the result obtained by CSIIRR achieves the highest correlation value for each Fourier frequency compared with the ones of WBP, IIRR and ICON. For the data with SNR=2 (Fig. 6c), the result obtained by CSIIRR achieves the highest correlation value for high frequency compared with the ones of WBP, IIRR and ICON.
Performance on real data
The proposed CSIIRR was further challenged by realworld datasets. We first made a whole tilt series reconstruction to see the behavior differences between the proposed CSIIRR and the other classical methods (e.g., IIRR and ICON). Then, a leaveoneout analysis was carried out to further make a quantitative analysis.
The leaveoneout [19] validation can show how precise the missing information is restored by a reconstruction method. In detail, we exclude the projection with the minimum tilt angle from the tilt series and then reconstruct the tomogram without considering the excluded projection. After the reconstruction, we reproject the tomogram according to the tilt angle value of the excluded projection and calculate the similarity between the reprojection and the exact excluded projection by Fourier ring correlation.
Experiment on centriole
The Centriole dataset is a tilt series of plastic embedded cell section around a centriole region, which was taken on a FEI TF30 microscope (operated at 300 kV) with a Gatan Camera. This dataset is downloaded from IMOD’s website [20]. The tilt angles of the projections range from +65.0^{∘} to −61.0^{∘} at 2^{∘} intervals. The size of each projection is 1024×1024 with a pixel size of 1.01 nm. The tilt series is aligned in advance by IMOD [20].
To reconstruct the tomogram, the number of iterations is set to 15 in IIRR. For the sake of fairness, the number of outerloop iterations in our proposed algorithm is also set to 15. The number of iterations for modified MP in CSIIRR is set to 3, and the number of atoms selected in each iterations M is set to 19,191,398 in modified MP. The iteration is set to ’30,120,300’ in ICON.
Figure 7 shows the XYslice and XZslice of the reconstructed tomograms by different methods. From Figs. 7a, b and e, f, it can be found that the tomogram reconstructed by CSIIRR has a clearer background comparing with the ones reconstructed by IIRR. Additionally, according to Fig. 7c, d, stripe noise appears around the boundary of the XYslice of tomogram from ICON while this kind of stripe noise does not appear around the boundary of the XYslice of tomogram from CSIIRR.
Figure 8 demonstrates the result of reprojection analysis. It can be found that the reprojection produced by IIRR is blurred while the reprojection produced by CSIIRR and ICON are very clear and keep high similarity with the exact excluded groundtruth. The Fourier ring correlation (FRC) curve (Fig. 8b) further indicates that CSIIRR could estimate the missing information more precisely compared with ICON.
Experiment on mitochondria
The Mitochondria dataset is a tilt series of mitochondria from mouse hepatic taken with a FEI Tecnai 20, with voltage at 200 kV. cells without fiducial markers. The tilt angles of the projections range from −52.0^{∘} to +59.0^{∘} at 1^{∘} intervals. The tilt series is aligned in advance by the markerfree alignment module in AuTom [16, 21].
Figure 9 shows the XYslice and XZslice of the reconstructed tomograms. As shown in Fig. 9a, b and e, f, the tomogram reconstructed by CSIIRR has a better contrast comparing with the ones reconstructed by IIRR. According to Fig. 9c, d, stripe noise appears around the boundary of the XYslice of tomogram from ICON while this kind of stripe noise does not appear around the boundary of the XYslice of tomogram from CSIIRR.
Figure 10 demonstrates the experiment result. As shown in Fig. 10a, the reprojection produced by IIRR has been blurred compared with the groundtruth. The reprojection produced by ICON looks better than IIRR. Nevertheless, strip artifacts still exist around the boundary of the projection produced by ICON. The reprojection produced by CSIIRR has the most visual details and is the most similar to the original excluded projection. The FRC curves shown in Fig. 10b further support our conclusion.
Conclusion
In this work, we proposed a novel algorithm for the reconstruction of electron tomography. By combining the concept of compressed sensing (CS) and improved iterative reconstructionreprojection (IIRR) together, this algorithm is able to suppress the effect of the missing wedge and restore the missing information.
Setting a proper value to the parameters in CSIIRR is really important to the performance of this method. First, the factor λ should be set carefully to ensure the convergence of the method. The author proposing the IIRR suggested that λ should be close to 1, so we set λ=0.99 in the experiment. Next, the number of atoms M should be set carefully to trade off the convergence rate and the sparsity of the reconstructed volume.
The proposed algorithm was challenged by both the simulated data and realworld data. The results show that the proposed algorithm has an obvious advance in the suppression of missing wedge effects and the restoration of missing information.
Availability of data and materials
The simulated data were generated from structures at the PDBj (Protein Data Bank Japan) (https://pdbj.org/emnavi/quick.php?id=emdb3489). The micrograph data are available from the download page of the website of bioinformatics and green network laboratory (http://ear.ict.ac.cn/). Our software implementation is freely available at CSTCloud (https://pan.cstcloud.cn/web/share.html?hash=CQBhG0ViQI8).
Abbreviations
 3D:

Threedimensional
 SNR:

Signaltonoise ratio
 FBP:

Filtered backprojection
 TV:

Total variation
 IRRTV:

Iterative reconstructionreprojection (IRR) algorithm with total variation (TV) constraint
 DIRRLF:

Discrete IRR algorithm and the finite impulse response (FIR) lowpass filter
 AIRR:

Algebraic iterative reconstructionreprojection
 IRRP:

Iterative reconstructionreprojection in projection space
 ICON:

Iterative compressedsensing optimized nonuniform fast fourier transform reconstruction
 CS:

Compressed sensing
 NUFFT:

Nonuniform fast fourier transform
 MP:

Matching pursuit
 CT:

Computed tomography
 MMP:

Modified MP
 PCC:

Pearson correlation coefficient
 SSIM:

Structural similarity
 FSC:

Fourier shell correlation
 FRC:

Fourier ring correlation
References
Frank J. Electron Tomography: Methods for Threedimensional Visualization of Structures in the Cell. New York: Springer; 2008.
Chen Y, Zhang Y, Zhang K, Deng Y, Wang S, Zhang F, Sun F. FIRT: Filtered iterative reconstruction technique with information restoration. J Struct Biol. 2016; 195(1):49–61.
Nassi M, Brody WR, Medoff BP, Macovski A. Iterative reconstructionreprojection: an algorithm for limited data cardiaccomputed tomography. IEEE Trans Biomed Eng. 1982; 5:333–41.
Medoff BP, Brody WR, Nassi M, Macovski A. Iterative convolution backprojection algorithms for image reconstruction from limited data. JOSA. 1983; 73(11):1493–500.
Kim J, Kwak K, Park S, Cho Z. Projection space iteration reconstructionreprojection. IEEE Trans Med Imaging. 1985; 4(3):139–43.
Wan X, He X, Zou W, Chen X, Cai M. A reconstruction algorithm with iterative reconstructionreprojection and an fir filter. In: Information Optics and Photonics Technology, vol. 5642. International Society for Optics and Photonics: 2005. p. 364–72. https://doi.org/10.1117/12.568775.
Duan X, Zhang L, Xing Y, Chen Z, Cheng J. Fewview projection reconstruction with an iterative reconstructionreprojection algorithm and tv constraint. IEEE Trans Nucl Sci. 2009; 56(3):1377–82.
Kim J, Kwak K, Park S. Iterative reconstructionreprojection in projection space. Proc IEEE. 1985; 73(6):1140–1.
Yazdanpanah AP, Regentova EE, Bebis G. Algebraic iterative reconstructionreprojection (airr) method for high performance sparseview ct reconstruction. Appl Math. 2016; 10(6):1–8.
Zhang Z, Xiong X, Yang W. An iterative image reconstruction algorithm with improved convergence. In: Proc. of 4th World Congress on Industrial Process Tomography. United Kingdom: VCIPT: 2005. p. 708–13.
Zhang Z, Zhang P. Improved iterative reprojection algorithm for tomographic image reconstruction. In: The 3rd International Symposium on Process Tomography. Poland: 2004. p. 183–6.
Zhang Z. Research on incomplete data tomography theory and application. PhD thesis. 2005.
Deng Y, Chen Y, Zhang Y, Wang S, Zhang F, Sun F. Icon: 3d reconstruction with ’missinginformation’restoration in biological electron tomography. J Struct Biol. 2016; 195(1):100–12.
Chen Y, Wang Z, Zhang J, Li L, Wan X, Sun F, Zhang F. Accelerating electron tomography reconstruction algorithm icon with gpu. Biophys Rep. 2017; 3(13):36–42.
Wang Z, Chen Y, Zhang J, Li L, Wan X, Liu Z, Sun F, Zhang F. Iconmic: Implementing a cpu/mic collaboration parallel framework for icon on tianhe2 supercomputer. J Comput Biol. 2018; 25(3):270–81.
Han R, Zhang F, Wan X, Fernández JJ, Sun F, Liu Z. A markerfree automatic alignment method based on scaleinvariant features. J Struct Biol. 2014; 186(1):167–80.
Han R, Wang L, Liu Z, Sun F, Zhang F. A novel fully automatic scheme for fiducial markerbased alignment in electron tomography. J Struct Biol. 2015; 192(3):403–17.
James NR, Brown A, Gordiyenko Y, Ramakrishnan V. Translational termination without a stop codon. Science. 2016; 354(6318):1437–40.
Cardone G, Grünewald K, Steven AC. A resolution criterion for electron tomography based on crossvalidation. J Struct Biol. 2005; 151(2):117–29.
Kremer JR, Mastronarde DN, McIntosh JR. Computer visualization of threedimensional image data using imod. J Struct Biol. 1996; 116(1):71–6.
Han R, Wan X, Wang Z, Hao Y, Zhang J, Chen Y, Gao X, Liu Z, Ren F, Sun F, et al.Autom: A novel automatic platform for electron tomography reconstruction. J Struct Biol. 2017; 199(3):196–208.
Acknowledgements
The authors thank Xiaohua Wan, Jingrong Zhang, Zihao Wang, Yu Hao, Hongjia Li for the valuable advice about this paper and thank Ying Mao for improving the language of this paper.
About this supplement
This article has been published as part of BMC Bioinformatics Volume 21 Supplement 6, 2020: Selected articles from the 15th International Symposium on Bioinformatics Research and Applications (ISBRA19): bioinformatics. The full contents of the supplement are available online at https://bmcbioinformatics.biomedcentral.com/articles/supplements/volume21supplement6.
Funding
Publication costs are funded by National Key Research and Development Program of China (No. 2017YFE0103900 and 2017YFA0504702), the Strategic Priority Research Program of the Chinese Academy of Sciences Grant (No. XDA19020400), the NSFC projects Grant (No. U1611263, U1611261, 61932018, and 61672493), Beijing Municipal Natural Science Foundation Grant (No. L182053) and Special Program for Applied Research on Super Computation of the NSFCGuangdong Joint Fund (the second phase).
Author information
Authors and Affiliations
Contributions
FZ and ZZ proposed the original idea about the combination of iterative reconstructionreprojection and compressed sensing. LL implemented the concept. RH designed the experiments. LL and RH processed the experiments. TG and ZL give some advice for the design of the optimization algorithm. LL, FZ, and RH drafted the manuscript. All authors read and approved the final manuscript.
Corresponding authors
Ethics declarations
Ethics approval and consent to participate
Not applicable.
Consent for publication
Not applicable.
Competing interests
The authors declare that they have no competing interests.
Additional information
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images or other third party material in this article are included in the article’s Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article’s Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated in a credit line to the data.
About this article
Cite this article
Li, L., Han, R., Zhang, Z. et al. Compressed sensing improved iterative reconstructionreprojection algorithm for electron tomography. BMC Bioinformatics 21 (Suppl 6), 202 (2020). https://doi.org/10.1186/s1285902035293
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/s1285902035293