 Research article
 Open Access
 Published:
Quadratic variance models for adaptively preprocessing SELDITOF mass spectrometry data
BMC Bioinformatics volume 11, Article number: 512 (2010)
Abstract
Background
Surface enhanced laser desorption/ionization timeofflight mass spectrometry (SELDI) is a proteomics tool for biomarker discovery and other high throughput applications. Previous studies have identified various areas for improvement in preprocessing algorithms used for protein peak detection. Bottomup approaches to preprocessing that emphasize modeling SELDI data acquisition are promising avenues of research to find the needed improvements in reproducibility.
Results
We studied the properties of the SELDI detector intensity response to matrix only runs. The intensity fluctuations and noise observed can be characterized by a natural exponential family with quadratic variance function (NEFQVF) class of distributions. These include as special cases many common distributions arising in practice (e.g. normal, Poisson). Taking this model into account, we present a modified AntoniadisSapatinas wavelet denoising algorithm as the core of our preprocessing program, implemented in MATLAB. The proposed preprocessing approach shows superior peak detection sensitivity compared to MassSpecWavelet for false discovery rate (FDR) values less than 25%.
Conclusions
The NEFQVF detector model requires that certain parameters be measured from matrix only spectra, leaving implications for new experiment design at the tradeoff of slightly increased cost. These additional measurements allow our preprocessing program to adapt to changing noise characteristics arising from intralaboratory and acrosslaboratory factors. With further development, this approach may lead to improved peak prediction reproducibility and nearly automated, high throughput preprocessing of SELDI data.
Background
Mass spectrometry is a promising technology for biomarker discovery [1]. There are a wide variety of mass spectrometers from which one could choose from during the design of a biomarker discovery experiment, reviewed in [2]. Matrix assisted laser desorption/ionization timeofflight mass spectrometry (MALDITOF MS, or just MALDI) can ionize whole proteins intact over a wide range of protein mass values, making it suitable for biomarker discovery in complex media such as blood serum, where both protein concentrations and masses vary greatly [3]. Surfaceenhanced laser desorption/ionization timeofflight mass spectrometry (SELDITOF MS, or just SELDI) [4] is a variant of MALDI that adds an onchip chromatographic separation step at the front end of the analysis pipeline. This, combined with robotautomated sample preparation, enables SELDI to be highthroughput, an attractive feature for many laboratories. For a recent review of the application of SELDI in the context of biomarker discovery, see [5].
The typical SELDI work flow involves the collection of samples (e.g. blood serum) from patients, application of the samples to SELDI ProteinChips^{®} selected for desired physicochemical properties, and analysis in the SELDI mass spectrometer. The raw data must be preprocessed to detect relevant peaks which correspond to proteins in the sample. Typical signal preprocessing steps performed are spectral alignment, denoising/smoothing, peak detection, peak matching, normalization, and quantification (see Figure 1 of [6]). The preprocessing of the raw SELDI spectra is typically accomplished using one of several available software packages (reviewed in [6–8]). Artifacts due to insufficient preprocessing of the data have, in the worst case, led to erroneous biological conclusions in early SELDI studies [9–11]. This fact inspired several important comparison studies of SELDI preprocessing algorithms [6–8, 12]. We now briefly summarize a few of the major contributions. For a more detailed overview, see the introduction of [6].
Coombes et al introduced the use of wavelets for denoising SELDI spectra [13], providing a more adaptive approach to denoise compared to moving average filters (e.g., as in [14]). Meanwhile, Morris et al introduced the notion of a mean spectrum, which represents average protein activity of a group of spectra. Under nonrestrictive assumptions, the mean spectrum has less noise and allows one to circumvent complicated peak matching algorithms that consolidate peak predictions among individual spectra into a consensus prediction. Malyarenko et al introduced a novel baseline removal algorithm based on a proposed charge accumulation model of the saturation phenomenon of the detector [15]. This was one of the first algorithms that was designed from the "bottomup", starting with physical considerations of SELDI. Later, deconvolution filters were shown to be a possible approach for improving mass resolution of SELDI [16–18].
Sköld et al analyzed singleshot spectra [19], the basic components of a final SELDI spectrum obtained by summing the results of many laser shots. They suggested that the observed counts in the single shot spectra may be proportional to a Poisson random variable, proposing a heteroscedastic model for the data. Meuleman et al also make use of singleshot spectra (subspectra) to derive a preprocessing algorithm based on analyzing these components separately [20].
In an attempt to improve on the bottomup approach to preprocessing, we analyze the statistics of the SELDI signal over a wide range of intensity values. Based on data presented herein, we propose a natural exponential family model with quadratic variance function for the statistics of the detector response for SELDI experiments. We believe this model is a plausible explanation for acquisition of singleshot spectra, summing of singleshot spectra into a final spectrum, and extracting protein estimates from a mean spectrum under a unified framework. Under this framework, we introduce a new preprocessing approach, adaptive to changing noise characteristics per spectrum and per experiment, and show favorable peak prediction performance.
Results
Bufferonly intensity measurements
Electronic measurements exhibit natural random fluctuations [21]. In many cases, these fluctuations are independent of the signal and are modeled as additive white Gaussian noise. In order to understand the nature of the noise fluctuations inherent to SELDI, we study the response of the detector under controlled experiments applying different buffers instead of protein samples under varying laser intensities (as in [22]). This eliminates the complexity introduced by adding serum to the chips while facilitating measurements of ion counts over a wide range of intensity values. In principle, this gives us a set of n repeated experiments from which we can study the statistics of the detector response compounded with noise and interference inherent to SELDI. In this fashion, we have generated two separate buffer + matrix datasets, denoted BUFFER1 and BUFFER2, which represent data generated on the same SELDI PBS IIc machine by different scientists and different machine parameters. BUFFER1/BUFFER2 contain 183/114 spectra, respectively.
We visualize all of the spectra in BUFFER1 and BUFFER2 in Figure 1. In particular, we are interested in analyzing the region between 3 and 30 kDa, since this is the mass focusing region in our experiments. In this region, the observations across spectra for a fixed time (mass) point represent approximately independent, identicallydistributed measurements in BUFFER1 or BUFFER2, respectively. Figure 1 shows the median, 75% quantile, and 25% quantile of BUFFER1 and BUFFER2. The median spectrum shows the form of an ordinary measurement, with any measurement between the 75% and 25% spectrum lines considered typical as well.
Figure 1 shows us the behavior of the typical buffer + baseline signal component seen in all SELDI raw spectra. Indeed, we see that changing different machine settings leads to different response properties. For BUFFER2, the median spectral response is large in the range shown, and the distribution of responses is symmetric about the median, whereas the distribution of detector response values for BUFFER1 are heavily skewed, and thus certainly not normally distributed.
We study the detector response (intensity output) for SELDI under varying input conditions, creating a detector response curve as follows. For each fixed time (mass) point across spectra from BUFFER1 in the mass focused region [3kDa; 30kDa], we estimate the mean intensity observed and the corresponding variance, with the same repeated for BUFFER2. These are displayed as a scatter plot in Figure 2 along with the best fit quadratic curve. Observing Figure 2 we see

1.
Intensity fluctuation/variance increases monotonically with the mean.

2.
The variance of the detector response is a quadratic function of the mean, to a very good approximation

3.
The detector response curves for BUFFER1 and BUFFER2 are quite different, and thus are dependent on the machine settings.
The detector response statistics thus exhibit a quadratic variance function. Briefly, a random variable X is said to have a quadratic variance function (QVF) if
with μ being the mean of X, V(μ) the variance, and υ_{ 0 }, υ_{1}, υ_{2} constants, some of which may be zero.
From these observations, summarized in Figures 1 and 2, it seems unlikely that an algorithm optimized for BUFFER1 would work well on BUFFER2 and vice versa. Further, neither a homoscedastic approach (e.g.  standard wavelet shrinkage [23]) or a simple heteroscedastic approach (e.g.  Poisson regression formulation [24]) to preprocessing the data is likely to be sufficient.
Data for evaluating preprocessing algorithms
We have generated two new datasets for evaluating preprocessing algorithms in order to improve upon purely simulationbased datasets used in previous comparison studies [6, 7]. A good comparison dataset should have the following properties (discussed previously in [6]):

1.
Exact protein content is known (and thus expectation of where "true" peaks will appear)

2.
Analyzed sample is complex containing many proteins/peaks

3.
Noise and baseline characteristics should be as close to those of real SELDI data as possible.
If one uses simulated data [6, 7, 25], complete control can be attained over requirements 1) and 2) at the expense of having noise/baseline characteristics that are overly ideal. If one uses purely real data, the noise, baseline, and artifacts that arise in actual experiments are present. However, this usually accompanies the tradeoff of either not knowing the exact protein content (e.g. complex serum data) or an overly simplified scenario (e.g.  spikein data).
We combine the advantages of purely simulated and real data by introducing the notion of a hybrid spectrum. To generate a hybrid spectrum, we use an implementation of the SimSpec 2.1 SELDI simulator [25, 26]http://bioinformatics.mdanderson.org/Software/Cromwell/simspec.zip to generate a "clean" SELDI spectrum, shown at the top of Figure 3. This gives an accurate peak shape characteristic as would be seen in low resolution SELDI/MALDI for given mass and ion abundance values, without any electronic noise or baseline present. We then select one of our buffer + matrix spectra (from either BUFFER1 or BUFFER2) and add the two together to produce the hybrid spectrum shown at the bottom of Figure 3. Thus, in a hybrid spectrum we know the exact virtual protein content specified to the simulator a priori while maintaining exactly the same noise, baseline, and other artifacts one encounters with real SELDI data.
Further details on the hybrid spectra can be found in the Methods section and in Additional file 1. The collection of hybrid spectra under different operating conditions results in test sets, denoted HYBRID1 and HYBRID2, with each test set containing thirty datasets of fifty hybrid spectra each. The mean performance of a preprocessing algorithm on HYBRID1 and HYBRID2 can be interpreted as the expected performance of the preprocessing approach in each separate operating condition in a repeated experiment or sampling from a homogeneous population (e.g.  cancer group or control group).
New preprocessing algorithms for SELDI
We have developed a set of MATLAB^{®} scripts for preprocessing SELDI spectra named LibSELDI. For information on how to obtain LibSELDI and the associated scripts used to produce the figures in this paper, contact the authors. We compare our preprocessing package to the MassSpecWavelet package from the Bioconductor project [27]. MassSpecWavelet has been established as one of the best approaches in terms of peak finding in recent comparison studies [6, 7], and has been downloaded > 6000 times in the past two years as of March 2010 http://bioconductor.org/packages/stats/bioc/MassSpecWavelet.html. Both packages have the advantage of having only one main useradjusted parameter.
In order to compare the performance of each preprocessing program, we generate operating characteristic curves (OC curves) [6, 20], one for each of the 30 datasets of HYBRID1 and HYBRID2, by varying the Peak Area threshold (LibSELDI) and signaltonoise ratio threshold (Snr.Th in MassSpecWavelet) parameters in the programs. Code snippets showing how MassSpecWavelet was tested can be found in Additional file 1. This allows us to understand the tradeoffs between false discovery rate (FDR) and sensitivity (TPR) achieved by each algorithm. The results for both the HYBRID1 and HYBRID2 collections are shown in Figure 4, where we have plotted the FDRaxis in log scale to emphasize the low FDR region which is usually of most interest in biomarker discovery applications. Note that, since both HYBRID1 and HYBRID2 are collections of datasets representing repeated trials (or equivalently a homogeneous population), the OC curves we show in Figure 4 are the mean OC curves across the 30 datasets for each.
The results show that LibSELDI tends to have a considerable advantage in the low FDR region, while MassSpecWavelet tends to have higher sensitivity for FDR > 25%. One way to summarize the performance of the algorithms is using the area under the OC curve for the FDR region of interest. We compute two area under the curve values, PAUC [6] (calculated for FDR ∈ [0, 50%]), and PAUC25 (calculated for FDR ∈ [0, 25%]). The results are shown in Table 1, where we have normalized each score separately so that a perfect PAUC25 (likewise, PAUC50) score is 100.
In Figure 5, we show the specific operating characteristics for LibSELDI and MassSpecWavelet for Dataset 2 of HYBRID1. While both algorithms perform well, LibSELDI resolves more than 90 proteins correctly before making a mistake. Since operating characteristics show false discovery rate along the xaxis rather than false positive rate (as in the traditional ROC curves), they tend to penalize more when false predictions are made with very few true proteins found. Indeed, in this case MassSpecWavelet got its first protein prediction correct but its second prediction wrong, leading to the point at FDR = 50%, TPR = 7%. Thus, operating characteristics with false discovery rate along the xaxis enforce the principle of conservative decision making, rewarding approaches that are successful with their initial large threshold (conservative) predictions and penalizing those that make mistakes early.
At FDR values greater than 30%, MassSpecWavelet outperforms LibSELDI. However, this is at the expense of generally more promiscuous predictions, since MassSpecWavelet generates 586 potential protein predictions compared to 250 for LibSELDI.
Discussion
We posit that the detector response is a member of the Natural Exponential Family with Quadratic Variance Function (NEFQVF), which is a proper subset of the exponential family of distributions [28]. Figures 1 and 2 show that assuming the detector response takes the form of a specific distribution is impractical, but that the detector response V(μ) has a QVF. The NEFQVF family of distributions occur often in practice and have the following useful properties, characterized by Morris [28]:

1.
If a random variable X ∈ NEFQVF, it is completely specified by its variance function V(μ)

2.
If X ∈ NEFQVF, a, b constants then aX + b is also NEFQVF

3.
Additivity: If X _{1}; X _{2} ∈ NEFQVF, then X _{1} + X _{2} is NEFQVF

4.
Affine combinations of normal, Poisson, gamma, binomial, negative binomial, and generalized hyperbolic secant distributed random variables generate all possible distributions in the NEFQVF family.
There are some physical reasons as to why the NEFQVF assumption could be reasonable as well. Some plausible justifications for the first two terms in Eq. (1) are:

1.
Constant Term: This is possibly due to thermal noise (additive Gaussian noise) which is common to all electronic measurement devices [21]

2.
Linear Term: The ability to detect an ion in a multiple stage electron multiplier, a common type of detector in MALDIlike instruments, is described by compound Poisson statistics [29].
The existence of a plausible physical explanation for the quadratic variance term remains an open question. However its effect is measured in both BUFFER1 and BUFFER2 and cannot be neglected. While the QVF model explains the data well in the mass focused region between 3 and 30 kDa, it is likely to break down at lower masses around 22.5 kDA where the baseline reaches a maximum. In this region the detector often saturates, introducing a nonlinearity into the data that we have not accounted for.
The success of our univariate model for SELDI may indicate that we have selected the most important feature to consider in the preprocessing of the data: namely, the fluctuations in the response of the ion detector subject to different inputs. The analysis of expression values of preprocessed data, on the other hand, requires multivariate methods as there are significant statistical dependencies between the peak heights corresponding to proteins that may be interacting. While these correlations are important in the analysis performed after the data is preprocessed, our results indicated it may be safe to ignore them during the preprocessing. While we have shown LibSELDI to be accurate for estimating peak m/z values, we have not assessed the usefulness approach for estimating peak intensities in this work. The utility of LibSELDI for accurately estimating peak intensities remains an open question and subject of future work.
It is entirely possible that the quadratic variance model could be applicable to other similar technologies such as MALDI and newer SELDI mass spectrometers. This, however, has not been confirmed.
Having buffer only spectra allows one to estimate the parameters of the detector response curve. Knowledge of the detector response curve enables us to apply the modified AntoniadisSapatinas denoising scheme described in the methods. Using this approach in our LibSELDI package yields excellent peak detection performance. We have proved this concept on HYBRID1 and HYBRID2 by estimating the QVF parameters of (1) using the bufferonly spectra that were randomly selected from BUFFER1 and BUFFER2 respectively. This implies that spots on SELDI chips should be reserved for bufferonly spectra. Thus, the tradeoff for using our approach is increased cost in terms of the number of chips one must use. The modified AntoniadisSapatinas denoising is computationally intensive as well, taking approximately seven minutes per spectrum on a highend workstation.
We argue that some of the cost is recovered by the potential for adaptive and accurate preprocessing, but not all. It may be possible to use QC and/or calibration samples to estimate the QVF as well rather than bufferonly spots. However, this would add in some additional variation due to the nature of the medium (serum, plasma, etc).
While LibSELDI outperforms MassSpecWavelet on the HYBRID1 and HYBRID2 test sets, the applicability of this comparison and of these results to purely real data remains an open question. There is some basic biological variability modeled in our test sets (see description in supplement of [6]). However, data from complex biological samples such as serum or plasma likely contains more biological variation and artifacts than we have modeled in HYBRID1 and HYBRID2. The investigation of how biological variation affects the model in QC samples is a work in progress.
In addition to achieving a better mean OC curve at lower FDR values, LibSELDI consistently predicts fewer peaks than MassSpecWavelet, leading to protein predictions closer to the true number of proteins in the data, as shown in Figure 6. This is further evidence that the adaptive modified AntoniadisSapatinas denoising approach using the NEFQVF model for the detector response is smoothing the spectra by close to the right amount.
Conclusions
We have shown that the variance of the intensity of a SELDI spectrum is quadratic in the mean signal strength. We further make the flexible assumption that the underlying distribution of the intensities is from a natural exponential family. From this point of view, we use a modified AntoniadisSapatinas wavelet shrinkage approach for denoising SELDI spectra. With this method at the core of our LibSELDI program for preprocessing SELDI data, we demonstrate excellent sensitivity at low false discovery rates. For applications that can tolerate higher false discovery rates, the MassSpecWavelet algorithm performs better in this region.
Our work has implications in the design of SELDI experiments. Namely, the modified AntoniadisSapatinas denoising technique performs well but requires an estimate of the quadratic variance function (QVF) describing the SELDI detector. This, in turn, is affected by machine settings. We have used bufferonly spectra to estimate the QVF. Thus, bufferonly spots could be interlaced on chips. We are investigating less expensive ways to estimate the QVF in future work.
Methods
Protocol for generating bufferonly spectra
Bufferonly spectra were generated by interspersing buffer only samples with protein samples from subjects (e.g. serum samples) and with pooled subject samples (for quality control) on the same chip. The bufferonly samples were spotted with wash buffer that was either PBS (phosphate buffered saline with various concentrations of phosphate and NaCl) based or acetonitrile + TFA (triflouroacetic acid) based, as manufacturer recommended per chip type. These buffer only samples were processed with the same washing steps as the subject samples, as described in [22], and then SPA matrix was applied to all spots.
The samples were analyzed with the Protein Biological System IIc™ SELDI mass spectrometer (Ciphergen Biosystems, Freemont, CA). The machine settings (e.g. laser intensity, detector sensitivity) and precise washing steps varied from buffer only spot to buffer only spot, and were generally different between BUFFER1 and BUFFER2. Note especially that laser intensities were generally higher for BUFFER2 than for BUFFER1. A detailed list of machine settings is given in the Additional file 1.
Hybrid data
Calculating performance statistics for comparison of MassSpecWavelet and LibSELDI requires a large number of spectra emulating an experiment that was repeated many times. To generate the HYBRID1 dataset, we combine each clean spectrum with one buffer+matrix spectrum from BUFFER1, and similarly we form HYBRID2 from BUFFER2 by combining those spectra with the same clean spectra.
A basic model of repetitive experiments for SELDI is available with SimSpec 2.1 that takes into account fluctuations in protein concentrations, m/z values, and prevalence in the data. Using the SimSpec 2.1 model developed at the MD Anderson Cancer Center [25, 26], we generate 30 datasets containing 50 clean (noise and matrixfree) spectra each. Each dataset consists of 150 virtual proteins and each spectrum within the given dataset contains a proper subset of these proteins with fluctuating parameters according to the model described in [25] and its supplement. The goal for the preprocessing programs in our performance evaluation is to reconstruct the master list of 150 virtual proteins characterizing the dataset. Repeated across all 30 datasets, we can calculate useful performance statistics. The properties of the 150 virtual proteins themselves are drawn from a prior distribution that was estimated from real data. See [25], or alternatively, the description in the supplement of [6].
We use sampling to overcome the limitation of having much fewer spectra in BUFFER1 and BUFFER2 than we have clean spectra in preparation for testing the algorithms. In principle the best way to construct the hybrid test sets would be to have one unique spectrum in BUFFER1 (likewise BUFFER2) for each spectrum in our clean proteinonly set. However, this would require 1500 buffer+matrix runs to be performed for both BUFFER1 and BUFFER2, an impractical amount of blank chips to run. Sampling from BUFFER1 (BUFFER2) provides a cost effective way to introduce variation in the noise/matrix characteristics between the datasets in HYBRID1 (HYBRID2).
Preprocessing the spectra
First we consider a model for a single SELDI spectrum, X(t). We observe X(t), a random process, on a discrete time grid t_{1},..., t_{ m }, where X(t) represents the intensity of the raw SELDI spectrum observed at time (equivalently mass) point t. For all t, we assume that X(t) is distributed according to a natural exponential family (NEF) with quadratic variance function (QVF) equal to V(μ(t)) as in Eq. (1). The variance function V(μ) completely characterizes the NEFQVF family. The goal of preprocessing in SELDI is to estimate μ(t), the expectation of X(t), which is the signal corresponding to ions that hit the detector. With a good estimate of μ(t), extracting peaks and estimating protein m/z values in a dataset is relatively straightforward.
As a side note we point out that a SELDI spectrum is actually a sum of single shot spectra. However, the additivity property of the NEFQVF family guarantees the sum is NEFQVF provided that the singleshot spectra are NEFQVF, agreeing with our detector response model and experimental observations.
Multiple spectra considerations
Rather than observe a single spectrum, the typical biomarker discovery approach is to generate at least one spectrum for each of n samples from an approximately homogeneous population. For example, one homogeneous population may be a group of early stage prostate cancer patients matched for age, race, etc. Assuming the samples are run on the same SELDI machine with the same operating conditions, we have
Our assumption that all n patients have the same underlying μ(t) is equivalent to assuming that the underlying biological condition being observed in each patient is approximately the same. Thus, we wish to estimate the underlying commonality μ(t) related to the biology of their condition expressed through the SELDI signal. We can mitigate some of the effects of the QVF by forming the mean spectrum (first introduced by [25]).
It is straightforward to show that
Thus, the mean spectrum concept is valuable under the assumptions of the NEFQVF model as well.
Modified AntoniadisSapatinas denoising
We now discuss estimation of μ(t) from the mean spectrum (3). Since the X_{ k }(t) are sampled on a discrete time grid (and thus X_{•}), we introduce vector notation
For any estimate \widehat{\mu}\left({x}_{\u2022}\right) of, μ we measure its fitness using the meansquarederror (MSE)
Antoniadis and Sapatinas proposed a wavelet shrinkage scheme to solve for \widehat{\mu} in (6) in the context of NEFQVF regression [30]. We summarize their main results. For our denoising, we use the orthogonal discrete wavelet transform with respect to the Symmlet 8 basis [31]. The transform can be represented by an m × m orthogonal matrix W,
Let h be a length m vector with entries taking values between 0 and 1. Let H = diag(h) be the m × m matrix defined by placing the entries of h along the main diagonal, all other entries 0. The class of estimators for \widehat{\mu}\left({x}_{\u2022}\right) considered by [30] take the form
This is the typical wavelet denoising scenario where each wavelet coefficient is left alone or shrunk towards zero according to some criterion, and is completely defined by the vector h. Antoniadis and Sapatinas showed that a good estimator for data from the NEFQVF family is given by choosing
The term {\widehat{\sigma}}^{2} is estimated as
Where V(x_{ • }) is the vector constructed by applying the QVF from (1) to each term of x_{ • }. (W · W ) is the matrix whose i, j element is the square of the i, j element of W. The parameters υ_{0}, υ_{1}, υ_{2} in (1) are measured from the bufferonly spectra, as described in the Results and Discussion section.
We make an intuitive modification to (9)
Thus our modified Antoniadis and Sapatinas estimator \tilde{h} uses {\widehat{\sigma}}^{2} in (8) rather than {\widehat{\sigma}}^{2}. The modification was introduced to account for cases when (9) may underestimate the noise when low amounts of observed signal are detected. Define
Then, our modified AntoniadisSapatinas estimate of μ is defined as
Peak detection/baseline removal
We consolidate the two preprocessing steps of baseline removal and peak detection typically performed separately into a single step as follows. We assume that the underlying μ(t) shown in (4) is the superposition of protein ions, s(t), and energyabsorbing matrix ions, b(t) striking the detector. It is well known that the distribution of the isotopes in our analyte of interest gives rise to a roughly Gaussian peak shape. Thus, we propose
where, {\mathcal{G}}_{\alpha}({t}_{j},{\sigma}_{j}) denotes a Gaussian kernel function centered at t_{ j }with standard deviation σ_{ j }and zero outside the interval [t_{ j } α, t_{ j }+ α].
Typically, s(t) is very sparse in the sense that it is mostly zero over the domain of the observed signal. Therefore, the local minima of our estimated baseline + noise signal \tilde{\mu} are points we may assume touch the baseline. From this point of view, once we have detected all the local minima in \tilde{\mu}, the baseline curve estimation problem reduces to an interpolation problem amongst these points. We have found through experimentation that piecewise cubic Hermite interpolating polynomials [32] are excellent interpolation functions.
The minima and maxima in \tilde{\mu} are found in one pass using the extrema function downloadable from MATLAB^{®} central file exchange. The maxima are the peaks in the mean spectrum potentially indicating proteins represented in our sample population while the minima correspond to samples from the baseline signal.
Each detected peak is quantified using peak area and a threshold is chosen based on the peak area measurement to generate the final prediction set.
Operating characteristics
The peaks we detect in \tilde{\mu} represent the initial set from which we choose our final estimates of proteins that are active in the population of interest. The choice of final estimate is accomplished using a peak area threshold (LibSELDI) or signaltonoise ratio measurement (Snr.Th in MassSpecWavelet). From each prediction, we calculate the observed false discovery rate (FDR) and true positive rate (TPR, also called sensitivity)
Where TP (the number of true positives) is the number of the 150 virtual protein m/z values having at least one predicted m/z value within 0.3% relative error. The FP is defined as the number of predicted m/z values not within 0.3% of any of the 150 virtual protein m/z values for this dataset. Similarly, FN is the number of the 150 virtual protein values without any predicted m/z value within 0.3% relative error.
For each dataset, a curve is fit to the operating points. Each operating curve is averaged to produce a mean operating characteristic, as shown in Figure 4. From this curve, the calculation of the areaunderthe curve is straightforward. For more details, see sections 2.2 and 2.2.1 of [6].
References
Hüttenhain R, Malmström J, Picotti P, Aebersold R: Perspectives of targeted mass spectrometry for protein biomarker verification. Curr Opin Chem Biol 2009, 13(5–6):518–525. 10.1016/j.cbpa.2009.09.014
Domon B, Aebersold R: Mass spectrometry and protein analysis. Science 2006, 312(5771):212–217. 10.1126/science.1124619
Corthals GL, Wasinger VC, Hochstrasser DF, Sanchez JC: The dynamic range of protein expression: a challenge for proteomic research. Electrophoresis 2000, 21(6):1104–1115. 10.1002/(SICI)15222683(20000401)21:6<1104::AIDELPS1104>3.0.CO;2C
Hutchens TW, Yip T: New Desorption Strategies for the Mass Spectrometric Analysis of Macromolecules. Rapid Commun Mass Spectrom 1993, 7(7):576–580. 10.1002/rcm.1290070703
Bock MD, de Seny D, Meuwis MA, Chapelle JP, Louis E, Malaise M, Merville MP, Fillet M: Challenges for biomarker discovery in body fluids using SELDITOFMS. J Biomed Biotechnol 2010, 2010: 906082.
Emanuele VA, Gurbaxani BM: Benchmarking currently available SELDITOF MS preprocessing techniques. Proteomics 2009, 9(7):1754–1762. 10.1002/pmic.200701171
CruzMarcelo A, Guerra R, Vannucci M, Li Y, Lau CC, Man TK: Comparison of algorithms for preprocessing of SELDITOF mass spectrometry data. Bioinformatics 2008, 24(19):2129–2136. 10.1093/bioinformatics/btn398
Wegdam W, Moerland PD, Buist MR, van Themaat EVL, Bleijlevens B, Hoefsloot HCJ, de Koster CG, Aerts JMFG: Classificationbased comparison of preprocessing methods for interpretation of mass spectrometry generated clinical datasets. Proteome Sci 2009, 7: 19. 10.1186/14775956719
Baggerly KA, Morris JS, Coombes KR: Reproducibility of SELDITOF protein patterns in serum: comparing datasets from different experiments. Bioinformatics 2004, 20(5):777–785. 10.1093/bioinformatics/btg484
Sorace JM, Zhan M: A data review and reassessment of ovarian cancer serum proteomic profiling. BMC Bioinformatics 2003, 4: 24. 10.1186/14712105424
Diamandis EP: Analysis of serum proteomic patterns for early cancer diagnosis: drawing attention to potential problems. J Natl Cancer Inst 2004, 96(5):353–356. 10.1093/jnci/djh056
Meuleman W, Engwegen JY, Gast MCW, Beijnen JH, Reinders MJ, Wessels LF: Comparison of normalisation methods for surfaceenhanced laser desorption and ionisation (SELDI) timeofflight (TOF) mass spectrometry data. BMC Bioinformatics 2008, 9: 88. 10.1186/14712105988
Coombes KR, Tsavachidis S, Morris JS, Baggerly KA, Hung MC, Kuerer HM: Improved peak detection and quantification of mass spectrometry data acquired from surfaceenhanced laser desorption and ionization by denoising spectra with the undecimated discrete wavelet transform. Proteomics 2005, 5(16):4107–4117. 10.1002/pmic.200401261
Fung ET, Enderwick C: ProteinChip clinical proteomics: computational challenges and solutions. BioTechniques 2002, Suppl: 34–38. 40–41 4041
Malyarenko DI, Cooke WE, Adam BL, Malik G, Chen H, Tracy ER, Trosset MW, Sasinowski M, Semmes OJ, Manos DM: Enhancement of sensitivity and resolution of surfaceenhanced laser desorption/ionization timeofflight mass spectrometric records for serum peptides using timeseries analysis techniques. Clin Chem 2005, 51: 65–74. 10.1373/clinchem.2004.037283
Malyarenko D, Cooke W, Tracy E, Trosset M, Semmes O, Sasinowski M, Manos D: Deconvolution filters to enhance resolution of dense timeofflight survey spectra in the timelag optimization range. Rapid Commun Mass Spectrom 2006, 20(11):1661–9. [Deconvolution filters; dense timeofflight survey spectral resolution;timelag optimization range;timedomain filters; timeofflight mass spectrometry signals;signaltonoise ratio; filtering;smoothing;time series method;matrixassisted laser desorption/ionization;biomolecules;nonlinear filters;] [Deconvolution filters; dense timeofflight survey spectral resolution;timelag optimization range;timedomain filters; timeofflight mass spectrometry signals;signaltonoise ratio; filtering;smoothing;time series method;matrixassisted laser desorption/ionization;biomolecules;nonlinear filters;] 10.1002/rcm.2487
Malyarenko DI, Cooke WE, Tracy ER, Drake RR, Shin S, Semmes OJ, Sasinowski M, Manos DM: Resampling and deconvolution of linear timeofflight records for enhanced protein profiling. Rapid Commun Mass Spectrom 2006, 20(11):1670–1678. 10.1002/rcm.2496
GatlinBunai CL, Cazares LH, Cooke WE, Semmes OJ, Malyarenko DI: Optimization of MALDITOF MS detection for enhanced sensitivity of affinitycaptured proteins spanning a 100 kDa mass range. J Proteome Res 2007, 6(11):4517–4524. 10.1021/pr0703526
Sköld M, Rydén T, Samuelsson V, Bratt C, Ekblad L, Olsson H, Baldetorp B: Regression analysis and modelling of data acquisition for SELDITOF mass spectrometry. Bioinformatics 2007, 23(11):1401–1409. 10.1093/bioinformatics/btm104
Meuleman W, Engwegen JYMN, Gast MCW, Wessels LFA, Reinders MJT: Analysis of mass spectrometry data using subspectra. BMC Bioinformatics 2009, 10(Suppl 1):S51. 10.1186/1471210510S1S51
van der Ziel A: Noise in Measurements. WileyInterscience; 1976.
Rollin D, Whistler T, Vernon SD: Laboratory methods to improve SELDI peak detection and quantitation. Proteome Sci 2007, 5: 9. 10.1186/1477595659
Donoho DL, Johnstone IM: Ideal spatial adaptation by wavelet shrinkage. Biometrika 1994, 81: 425–455. 10.1093/biomet/81.3.425
Donoho DL: Nonlinear Wavelet Methods for Recovery of Signals, Densities, and Spectra from Indirect and Noisy Data. In In Proceedings of Symposia in Applied Mathematics. American Mathematical Society; 1993:173–205.
Morris JS, Coombes KR, Koomen J, Baggerly KA, Kobayashi R: Feature extraction and quantification for mass spectrometry in biomedical applications using the mean spectrum. Bioinformatics 2005, 21(9):1764–1775. 10.1093/bioinformatics/bti254
Coombes KR, Koomen JM, Baggerly KA, Morris JS, Kobayashi R: Understanding the characteristics of mass spectrometry data through the use of simulation. Cancer Informatics 2005, 1: 41–52.
Du P, Kibbe WA, Lin SM: Improved peak detection in mass spectrum by incorporating continuous wavelet transformbased pattern matching. Bioinformatics 2006, 22(17):2059–2065. 10.1093/bioinformatics/btl355
Morris CN: Natural exponential families with quadratic variance functions. The Annals of Statistics 1982, 10: 65–80. 10.1214/aos/1176345690
Dietz LA: Basic Properties of Electron Multiplier Ion Detection and Pulse Counting Methods in Mass Spectrometry. Rev Sci Instrum 1965, 36(12):1763–1770. 10.1063/1.1719460
Antoniadis A, Sapatinas T: Wavelet shrinkage for natural exponential families with quadratic variance functions. Biometrika 2001, 88(3):805–820. 10.1093/biomet/88.3.805
Daubechies I: Ten lectures on wavelets. Philadelphia, PA, USA: Society for Industrial and Applied Mathematics; 1992.
Fritsch FN, Carlson RE: Monotone Piecewise Cubic Interpolation. SIAM j Numerical Analysis 1980, 17: 238–246. 10.1137/0717021
Acknowledgements
The authors would like to thank Dr. Gitika Panicker and Dr. Dominique Rollin of the Centers for Disease Control and Prevention, Atlanta, GA, USA for conception, generation, and acquisition of the buffer+matrix data used for BUFFER1 and BUFFER2, respectively. The authors would also like to thank Dr. Sally Lin and Dr. Toni Whistler (also at the CDC) for useful discussions that led to improvements in the manuscript. Lastly, we would to thank Dr. Elizabeth Unger and Dr. William C. Reeves of the CDC and Dr. G. Tong Zhou of Georgia Tech for support during the research and writing of the manuscript.
This work was supported in part by NCI's Early Detection Research Network through IAA Y1CN500501 and Y1CN010101.
This research was supported in part by an appointment to the Research Participation Program at the Centers for Disease Control and Prevention, National Center for Emerging and Zoonotic Infectious Diseases, Division of HighConsequence Pathogens and Pathology, Chronic Viral Diseases Branch, administered by the Oak Ridge Institute for Science and Education through an interagency agreement between the U.S. Department of Energy and the CDC.
The findings and conclusions in this report are those of the authors and do not necessarily represent the views of the funding agencies. The authors declare no conflict of interest.
Author information
Authors and Affiliations
Corresponding author
Additional information
Authors' contributions
VE and BG conceived and developed the theoretical aspects of the study, analyzed the results, and wrote the manuscript. VE developed LibSELDI and ran the simulations. Both authors have read and approve of the manuscript.
Electronic supplementary material
12859_2010_4095_MOESM1_ESM.PDF
Additional file 1: Experiment and simulation settings. This file contains additional details about how simulations and experiments were carried out. (PDF 82 KB)
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Emanuele, V.A., Gurbaxani, B.M. Quadratic variance models for adaptively preprocessing SELDITOF mass spectrometry data. BMC Bioinformatics 11, 512 (2010). https://doi.org/10.1186/1471210511512
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/1471210511512
Keywords
 False Discovery Rate
 Detector Response
 Machine Setting
 Wavelet Shrinkage
 Natural Exponential Family